Cardiovascular state determination apparatus and method of use thereof

ABSTRACT

The invention comprises a method and apparatus for estimating state of a cardiovascular system, comprising a cardiac stroke volume analyzer, comprising: (1) a blood pressure sensor generating a time-varying pressure state waveform output from a limb of the person; (2) a system processor connected to the blood pressure sensor; and (3) a dynamic state-space model; the system processor receiving cardiovascular input data, from the blood pressure sensor, related to a transient pressure state of the cardiovascular system; at least one probabilistic model, of the dynamic state-space model, operating on the time-varying pressure state waveform output to generate a probability distribution function to a non-pressure state of the cardiovascular system; iteratively updating the probability distribution function using output from the blood pressure sensor; and processing the probability distribution function to generate a non-pressure state output related to stroke volume of a heart of the person and arterial compliance of the person.

CROSS-REFERENCES TO RELATED APPLICATIONS

This application is a continuation-in-part of U.S. patent application Ser. No. 15/251,779 filed Aug. 30, 2016 now U.S. Pat. No. 10,460,843, which is a continuation-in-part of U.S. patent application Ser. No. 13/181,140 filed Jul. 12, 2011 now U.S. Pat. No. 9,451,886, which is a continuation-in-part of U.S. patent application Ser. No. 13/181,027, filed Jul. 12, 2011 now U.S. Pat. No. 8,494,829, which: is a continuation-in-part of U.S. patent application Ser. No. 12/796,512, filed Jun. 8, 2010 now U.S. Pat. No. 9,060,722, which is a continuation-in-part of U.S. patent application Ser. No. 12/640,278, filed Dec. 17, 2009 now abandoned, which claims benefit of U.S. provisional patent application No. 61/171,802, filed Apr. 22, 2009; claims benefit of U.S. provisional patent application No. 61/366,437 filed Jul. 21, 2010; claims benefit of U.S. provisional patent application No. 61/372,190 filed Aug. 10, 2010; and claims benefit of U.S. provisional patent application No. 61/373,809 filed Aug. 14, 2010, all of which are incorporated herein in their entirety by this reference thereto.

STATEMENT REGARDING FEDERALLY SPONSORED RESEARCH OR DEVELOPMENT

The U.S. Government may have certain rights to this invention pursuant to Contract Number IIP-0839734 awarded by the National Science Foundation.

BACKGROUND OF THE INVENTION Field of the Invention

The present invention relates generally to apparatus and methods for processing and/or representing sensor data, such as mechanical or medical sensor data.

Discussion of the Related Art

Mechanical devices and biomedical monitoring devices such as pulse oximeters, glucose sensors, electrocardiograms, capnometers, fetal monitors, electromyograms, electroencephalograms, and ultrasounds are sensitive to noise and artifacts. Typical sources of noise and artifacts include baseline wander, electrode-motion artifacts, physiological artifacts, high-frequency noise, and external interference. Some artifacts can resemble real processes, such as ectopic beats, and cannot be removed reliably by simple filters; however, these are removable by the techniques taught herein. In addition, mechanical devices and biomedical monitoring devices address a limited number of parameters. It would be desirable to expand the number of parameters measured, such as to additional biomedical state parameters.

Patents related to the current invention are summarized herein.

Mechanical Systems

Several reports of diagnostics and prognostics applied to mechanical systems have been reported.

Vibrational Analysis

R. Klein “Method and System for Diagnostics and Prognostics of a Mechanical System”, U.S. Pat. No. 7,027,953 B2 (Apr. 11, 2006) describes a vibrational analysis system for diagnosis of health of a mechanical system by reference to vibration signature data from multiple domains, which aggregates several features applicable to a desired fault for trend analysis of the health of the mechanical system.

Intelligent System

S. Patel, et. al. “Process and System for Developing Predictive Diagnostic Algorithms in a Machine”, U.S. Pat. No. 6,405,108 B1 (Jun. 11, 2002) describe a process for developing an algorithm for predicting failures in a system, such as a locomotive, comprising conducting a failure mode analysis to identify a subsystem, collecting expert data on the subsystem, and generating a predicting signal for identifying failure modes, where the system uses external variables that affect the predictive accuracy of the system.

C. Bjornson, “Apparatus and Method for Monitoring and Maintaining Plant Equipment”, U.S. Pat. No. 6,505,145 B1 (Jan. 11, 2003) describes a computer system that implements a process for gathering, synthesizing, and analyzing data related to a pump and/or a seal, in which data are gathered, the data is synthesized and analyzed, a root cause is determined, and the system suggests a corrective action.

C. Bjornson, “Apparatus and Method for Monitoring and Maintaining Plant Equipment”, U.S. Pat. No. 6,728,660 B2 (Apr. 27, 2004) describes a computer system that implements a process for gathering, synthesizing, and analyzing data related to a pump and/or a seal, in which data are gathered, the data is synthesized and analyzed, and a root cause is determined to allow a non-specialist to properly identify and diagnose a failure associated with a mechanical seal and pump.

K. Pattipatti, et. al. “Intelligent Model-Based Diagnostics for System Monitoring, Diagnosis and Maintenance”, U.S. Pat. No. 7,536,277 B2 (May 19, 2009) and K. Pattipatti, et. al. “Intelligent Model-Based Diagnostics for System Monitoring, Diagnosis and Maintenance”, U.S. Pat. No. 7,260,501 B2 (Aug. 21, 2007) both describe systems and methods for monitoring, diagnosing, and for condition-based maintenance of a mechanical system, where model-based diagnostic methodologies combine or integrate analytical models and graph-based dependency models to enhance diagnostic performance.

Inferred Data

R. Tryon, et. al. “Method and Apparatus for Predicting Failure in a System”, U.S. Pat. No. 7,006,947 B2 (Feb. 28, 2006) describe a method and apparatus for predicting system failure or reliability using a computer implemented model relying on probabilistic analysis, where the model uses data obtained from references and data inferred from acquired data. More specifically, the method and apparatus uses a pre-selected probabilistic model operating on a specific load to the system while the system is under operation.

Virtual Prototypinq

R. Tryon, et. al. “Method and Apparatus for Predicting Failure of a Component”, U.S. Pat. No. 7,016,825 B1 (Mar. 21, 2006) describe a method and apparatus for predicting component failure using a probabilistic model of a material's microstructural-based response to fatigue using virtual prototyping, where the virtual prototyping simulates grain size, grain orientation, and micro-applied stress in fatigue of the component.

R. Tryon, et. al. “Method and Apparatus for Predicting Failure of a Component, and for Determining a Grain Orientation Factor for a Material”, U.S. Pat. No. 7,480,601 B2 (Jan. 20, 2009) describe a method and apparatus for predicting component failure using a probabilistic model of a material's microstructural-based response to fatigue using a computer simulation of multiple incarnations of real material behavior or virtual prototyping.

Medical Systems

Several reports of systems applied to biomedical systems have been reported.

Lung Volume

M. Sackner, et. al. “Systems and Methods for Respiratory Event Detection”, U.S. patent application no. 2008/0082018 A1 (Apr. 3, 2008) describe a system and method of processing respiratory signals from inductive plethysmographic sensors in an ambulatory setting that filters for artifact rejection to improve calibration of sensor data and to produce output indicative of lung volume.

Pulse Oximeter

J. Scharf, et. al. “Separating Motion from Cardiac Signals Using Second Order Derivative of the Photo-Plethysmograph and Fast Fourier Transforms”, U.S. Pat. No. 7,020,507 B2 (Mar. 28, 2006) describes the use of filtering photo-plethysmograph data in the time domain to remove motion artifacts.

M. Diab, et. al. “Plethysmograph Pulse Recognition Processor”, U.S. Pat. No. 6,463,311 B1 (Oct. 8, 2002) describe an intelligent, rule-based processor for recognition of individual pulses in a pulse oximeter-derived photo-plethysmograph waveform operating using a first phase to detect candidate pulses and a second phase applying a plethysmograph model to the candidate pulses resulting in period and signal strength of each pulse along with pulse density.

C. Baker, et. al. “Method and Apparatus for Estimating Physiological Parameters Using Model-Based Adaptive Filtering”, U.S. Pat. No. 5,853,364 (Dec. 29, 1998) describe a method and apparatus for processing pulse oximeter data taking into account physical limitations using mathematical models to estimate physiological parameters.

Cardiac

J. McNames, et. al. “Method, System, and Apparatus for Cardiovascular Signal Analysis, Modeling, and Monitoring”, U.S. patent application publication no. 2009/0069647 A1 (Mar. 12, 2009) describe a method and apparatus to monitor arterial blood pressure, pulse oximetry, and intracranial pressure to yield heart rate, respiratory rate, and pulse pressure variation using a statistical state-space model of cardiovascular signals and a generalized Kalman filter to simultaneously estimate and track the cardiovascular parameters of interest.

M. Sackner, et. al. “Method and System for Extracting Cardiac Parameters from Plethysmograph Signals”, U.S. patent application publication no. 2008/0027341 A1 (Jan. 31, 2008) describe a method and system for extracting cardiac parameters from ambulatory plethysmographic signal to determine ventricular wall motion.

Hemorrhage

P. Cox, et. al. “Methods and Systems for Non-Invasive Internal Hemorrhage Detection”, International Publication no. WO 2008/055173 A2 (May 8, 2008) describe a method and system for detecting internal hemorrhaging using a probabilistic network operating on data from an electrocardiogram, a photoplethysmogram, and oxygen, respiratory, skin temperature, and blood pressure measurements to determine if the person has internal hemorrhaging.

Disease Detection

V. Karlov, et. al. “Diagnosing Inapparent Diseases From Common Clinical Tests Using Bayesian Analysis”, U.S. patent application publication no. 2009/0024332 A1 (Jan. 22, 2009) describe a system and method of diagnosing or screening for diseases using a Bayesian probability estimation technique on a database of clinical data.

Statement of the Problem

Mechanical and biomedical sensors are typically influenced by multiple sources of contaminating signals that often overlap the frequency of the signal of interest, making it difficult, if not impossible, to apply conventional filtering. Severe artifacts such as occasional signal dropouts due to sensor movement or large periodic artifacts are also difficult to filter in real time. Biological sensor hardware can be equipped with a computer comprising software for post-processing data and reducing or rejecting noise and artifacts. Current filtering techniques typically use some knowledge of the expected frequencies of interest where the sought-after physiological information should be found.

Adaptive filtering has been used to attenuate artifacts in pulse oximeter signals corrupted with overlapping frequency noise bands by estimating the magnitude of noise caused by patient motion and other artifacts and canceling its contribution from pulse oximeter signals during patient movement. Such a time correlation method relies on a series of assumptions and approximations to the expected signal, noise, and artifact spectra, which compromises accuracy, reliability, and general applicability.

Filtering techniques based on Kalman and extended Kalman techniques offer advantages over conventional methods and work well for filtering linear systems or systems with small nonlinearities and Gaussian noise. These filters, however, are not adequate for filtering highly nonlinear systems and non-Gaussian/non-stationary noise. Therefore, obtaining reliable biomedical signals continue to present problems, particularly when measurements are made in mobile, ambulatory, and physically active patients.

Existing data processing techniques, including adaptive noise cancellation filters, are unable to extract information that is hidden or embedded in biomedical signals and also discard some potentially valuable information.

Existing medical sensors sense a narrow spectrum of medical parameters and states. What is needed is a system readily expanding the number of biomedical states determined.

A method or apparatus for extracting additional useful information from a mechanical sensor in a mechanical system, a biomedical system, and/or a system component or sub-component is needed to provide users additional and/or clearer information.

SUMMARY OF THE INVENTION

The invention comprises use of fused data in a probabilistic model to extract, filter, estimate and/or add additional information about a system based on data from a sensor.

DESCRIPTION OF THE FIGURES

A more complete understanding of the present invention is derived by referring to the detailed description and claims when considered in connection with the Figures, wherein like reference numbers refer to similar items throughout the Figures.

FIG. 1 illustrates operation of the intelligent data extraction algorithm on a biomedical apparatus;

FIG. 2 provides a block diagram of a data processor;

FIG. 3 is a flow diagram of a probabilistic digital signal processor;

FIG. 4 illustrates a dual estimator;

FIG. 5 expands the dual estimator;

FIG. 6 illustrates state and model parameter estimators;

FIG. 7 provides inputs and internal operation of a dynamic state-space model;

FIG. 8 is a flow chart showing the components of a hemodynamics dynamic state-space model;

FIG. 9 is a chart showing input sensor data, FIG. 9A; processed output data of heart rate, FIG. 9B; stroke volume, FIG. 9C; cardiac output, FIG. 9D; oxygen, FIG. 9E; and pressure, FIG. 9F, from a data processor configured to process pulse oximetry data;

FIG. 10 is a chart showing input sensor data, FIG. 10A, and processed output data, FIGS. 10A-10E, from a data processor configured to process pulse oximetry data under a low blood perfusion condition;

FIG. 11 is a flow chart showing the components of a electrocardiograph dynamic state-space model;

FIG. 12 is a chart showing noisy non-stationary ECG sensor data input, FIG. 12A and FIG. 12B and processed heart rate and ECG output, FIG. 12A and FIG. 12B, for a data processor configured to process ECG sensor data;

FIG. 13A and FIG. 13B are charts showing input ECG sensor data and comparing output data from a data processor according to the present invention with output data generating using a Savitzky-Golay FIR data processing algorithm;

FIG. 14 illustrates fusion of data from multiple instruments;

FIG. 15 illustrates fusion of biomedical data, accelerometer data, and/or environmental data;

FIG. 16 shows integration of multiple data streams into a joint processor;

FIG. 17 illustrates a fusion dynamic state-space model;

FIG. 18 illustrates combination of medical data streams into a physics based model; and

FIG. 19 provides a flowchart of dynamic state-space model diagnostics used as prognosis and control.

DETAILED DESCRIPTION OF THE INVENTION

The invention comprises use of a method and apparatus for estimating state of a cardiovascular system of a person, comprising the steps of providing a cardiac stroke volume analyzer, comprising: (1) a blood pressure sensor, the blood pressure sensor generating a time-varying pressure state waveform output from a limb of the person; (2) a system processor connected to the blood pressure sensor; and (3) a dynamic state-space model; the system processor receiving cardiovascular input data, from the blood pressure sensor, related to a transient pressure state of the cardiovascular system; at least one probabilistic model, of the dynamic state-space model, operating on the time-varying pressure state waveform output to generate a probability distribution function to a non-pressure state of the cardiovascular system; iteratively updating the probability distribution function using synchronized updated time-varying pressure state waveform output from the blood pressure sensor; and the system processor processing the probability distribution function to generate a non-pressure state output related to at least one of a stroke volume of a heart of the person and arterial compliance of the person, the output displayed to at least one of the person and a doctor.

The system applies to the mechanical and medical fields. Herein, for clarity the system is applied to biomedical devices, though the system concepts apply to mechanical apparatus.

In one embodiment, an intelligent data extraction algorithm (IDEA) is used in a system, which combines a dynamic state-space model with a probabilistic digital signal processor to estimate a parameter, such as a biomedical parameter. Initial probability distribution functions are input to a dynamic state-space model, which iteratively operates on probability distribution functions (PDFs), such as state and model probability distribution functions, to generate a prior probability distribution function, which is input into a probabilistic updater. The probabilistic updater integrates sensor data with the prior probability distribution function to generate a posterior probability distribution function passed to a probabilistic sampler, which estimates one or more parameters using the posterior, which is output or re-sampled and used as an input to the dynamic state-space model in the iterative algorithm. In various embodiments, the probabilistic data signal processor is used to filter output and/or estimate a value of a new physiological parameter from a biomedical device using appropriate physical models, which optionally include biomedical, chemical, electrical, optical, mechanical, and/or fluid based models. For clarity, examples of heart and cardiovascular medical devices are provided.

In one example, an analyzer is configured to: (1) receive discrete first input data, related to a first sub-system of the system, from a first instrument and (2) receive discrete second input data, related to a second sub-system of the system, from a second instrument. The analyzer optionally includes a system processor configured to fuse the first input data and the second input data into fused data. The system processor optionally includes: (1) a probabilistic processor configured to convert the fused data into at least two probability distribution functions and (2) a dynamic state-space model, the dynamic state-space model including at least one probabilistic model configured to operate on the at least two probability distribution functions. The system processor iteratively circulates the at least two probability distribution functions in the dynamic state-space model in synchronization with receipt of at least one of: (1) updated first input data and (2) updated second input data. The system processor is further configured to process the probability distribution functions to generate an output related to the state of the system.

In another example, an analyzer is configured for processing sensor data representative of a body where the analyzer includes: a physical model representative of function of a body constituent; the physical model coded into a digital signal processor electrically connected, optionally wirelessley, to a computer embedded in the analyzer. The digital signal processor is configured to: (1) generate a prior probability distribution function using the physical model and (2) repetitively fuse input data originating from at least two types of medical instruments with the prior probability distribution function to generate a posterior probability distribution function. Further, the processor is configured to process the posterior probability distribution function to generate an output of at least one of: (1) a monitored parameter value representative of the body and (2) an estimated parameter value representative of the body.

In various embodiments, the probabilistic digital signal processor comprises one or more of a dynamic state-space model, a dual or joint updater, and/or a probabilistic sampler, which process input data, such as sensor data and generates an output. Preferably, the probabilistic digital signal processor (1) iteratively processes the data and/or (2) uses a mathematical model of the physical system in processing the input data.

The probabilistic digital signal processor optionally:

-   -   operates on or in conjunction with a sensor in a mechanical         system;     -   filters input data;     -   operates using data from a medical meter, where the medical         meter yields a first physical parameter from raw data, to         generate a second physical parameter not output by the medical         meter;     -   operates on discrete/non-probabilistic input data, such as from         a mechanical device or a medical device to generate a         probabilistic output function;     -   iteratively circulates or dynamically circulates a probability         distribution function through at least two of the dynamic         state-space model, the dual or joint updater, and/or the         probabilistic sampler;     -   fuses or combines output from multiple sensors, such as two or         more medical devices; and     -   prognosticates probability of future events.

To facilitate description of the probabilistic digital signal processor, a non-limiting example of a hemodynamics process model is provided. In this example, the probabilistic digital signal processor is provided:

-   -   raw sensor data, such as current, voltage, and/or resistance;         and/or     -   output from a medical device to a first physical or chemical         parameter.

In this example, the medical device is a pulse oximeter and the first parameter from the pulse oximeter provided as input to the probabilistic digital signal processor is one or more of:

-   -   raw data, such as a voltage waveform that correlates to light         absorption by blood;     -   heart rate; and/or     -   blood oxygen saturation.

The probabilistic digital signal processor uses a physical model, such as a probabilistic model, to operate on the first physical parameter to generate a second physical parameter, where the second physical parameter is not the first physical parameter. For example, the output of the probabilistic digital signal processor when provided with the pulse oximeter data is one or more of:

-   -   a heart stroke volume;     -   a cardiac output flow rate;     -   an aortic blood pressure; and/or     -   a radial blood pressure.

Optionally, the output from the probabilistic model is an updated, an error filtered, and/or a smoothed version of the original input data, such as a smoothed blood oxygen saturation percentage as a function of time. The hemodynamics model is further described, infra.

To facilitate description of the probabilistic digital signal processor, another non-limiting example of an electrocardiograph process model is provided. In this example, the probabilistic digital signal processor is provided:

-   -   raw sensor data, such as intensity, an electrical current,         and/or a voltage; and/or     -   output from a medical device, such as an electrocardiogram, to a         first physical or electrical parameter.

In this example, the medical device is a electrocardiograph and the first physical or electrical parameter from the electrocardiograph system provided as input to the probabilistic digital signal processor is one or more of:

-   -   raw data; and/or     -   an electrocardiogram.

The probabilistic digital signal processor uses a physical model, such as a probabilistic model, to operate on the first physical parameter to generate a second physical parameter or an indicator, where the second physical parameter is not the first physical parameter. For example, the output of the probabilistic digital signal processor when provided with the electrocardiogram or raw data is one or more of:

-   -   an arrhythmia detection;     -   an ischemia warning; and/or     -   a heart attack prediction.

Optionally, the output from the probabilistic model is an updated, error filtered, or smoothed version of the original input data. For example, the probabilistic processor uses a physical model where the output of the model processes low signal-to-noise ratio events to yield an early warning of any of the arrhythmia detection, the ischemia warning, and/or the heart attack prediction. The electrocardiograph model is further described, infra.

To still further facilitate description of the probabilistic digital signal processor, non-limiting fusion examples are provided, which combine data from one or more of:

-   -   a mechanical system;     -   a sensor monitoring a mechanical device;     -   an electrodynamics based medical device;     -   a hemodynamic based medical device;     -   accelerometer data; and     -   an environmental meter.

As further described, supra, fusion of signals or sensor data from a plurality of devices allows:

-   -   detection of a false positive or false negative signal from a         first device with a second device;     -   noise recognized in first sensor data as the noise is not         present in a second sensor type or is correlated with noise of         the second sensor type;     -   fusion of environmental data with medical data;     -   determination of an additional parameter not independently         measured with individual data types of the fused data;     -   electrocardiograph data to aid in analysis of pulse oximeter         data and vise-versa; and/or     -   electrodynamic information to aid in analysis of hemodynamic         information and vise-versa.         Deterministic vs. Probabilistic Models

Typically, computer-based systems use a mapping between observed symptoms of failure and the equipment where the mapping is built using deterministic techniques. The mapping typically takes the form of a look-up table, a symptom-problem matrix, trend analysis, and production rules. In stark contrast, alternatively probabilistic models are used to analyze a system. An example of a probabilistic model, referred to herein as an intelligent data extraction system is provided, infra.

Intelligent Data Extraction System

Referring now to FIG. 1, an algorithm based intelligent data extraction system 100 is illustrated. The intelligent data extraction system 100 uses a controller 110 to control a sensor 120. The sensor 120 is used to measure a parameter and/or is incorporated into a biomedical apparatus 130. Optionally, the controller 110 additionally controls the medical apparatus and/or is built into the biomedical apparatus 130. The sensor 120 provides readings to a data processor or a probabilistic digital signal processor 200, which provides feedback to the controller 110 and/or provides output 150. In one embodiment, the controller 110 comprises a microprocessor in a computer or computer system, an embedded device, and/or an embedded processor.

Herein, to enhance understanding and for clarity of presentation, non-limiting examples of an intelligent data extraction system operating on a hemodynamics biomedical devices are used to illustrate methods, systems, and apparatus described herein. Generally, the methods, systems, and apparatus described herein extend to any apparatus having a moveable part and/or to any medical device. Examples of the dynamic state-space model with a probabilistic digital signal processor used to estimate parameters of additional biomedical systems are provided after the details of the processing engine are presented.

Still referring to FIG. 1, in a pulse oximeter example the controller 110 controls a sensor 120 in the pulse oximeter apparatus 130. The sensor 120 provides readings, such as a spectral reading to the probabilistic digital signal processor 200, which is preferably a probability based data processor. The probabilistic digital signal processor 200 optionally operates on the input data or provides feedback to the controller 110, such as state of the patient, as part of a loop, iterative loop, time series analysis, and/or generates the output 150, such as a smoothed biomedical state parameter or a new biomedical state parameter. For clarity, the pulse oximeter apparatus is used repetitively herein as an example of the biomedical apparatus 130 upon which the intelligent data extraction system 100 operates. The probabilistic digital signal processor 200 is further described, infra.

Data Processor

Referring now to FIG. 2, the probabilistic digital signal processor 200 of the intelligent data extraction system 100 is further described. Generally, the data processor includes a dynamic state-space model 210 (DSSM) and a probabilistic updater 220 that iteratively or sequentially operates on sensor data 122 from the sensor 120. The probabilistic updater 220 outputs a probability distribution function to a parameter updater or a probabilistic sampler 230, which generates one or more parameters, such as an estimated diagnostic parameter, which is sent to the controller 110, is used as part of an iterative loop as input to the dynamic state-space model 210, and/or is a basis of the output 150. The dynamic state-space model 210 and probabilistic updater 220 are further described, infra.

Referring now to FIG. 3, the probabilistic digital signal processor 200 is further described. Generally, a probability function, a probability distribution function (PDF), an initial probability distribution function, or a set of initial probability distribution functions 310 are input to the dynamic state-space model 210. In a process 212, the dynamic state-space model 210 operates on the initial probability distribution functions 310 to generate a prior probability distribution function, hereinafter also referred to as a prior or as a prior PDF. For example, an initial state parameter 312 probability distribution function and an initial model parameter 314 probability distribution function are provided as initial inputs to the dynamic state-space model 210. The dynamic state-space model 210 operates on the initial state parameter 312 and/or initial model parameter 314 to generate the prior probability distribution function, which is input to the probabilistic updater 220. In a process 320, the probabilistic updater 220 integrates sensor data, such as timed sensor data 122, by operating on the sensor data and on the prior probability distribution function to generate a posterior probability distribution function, herein also referred to as a posterior or as a posterior PDF. In a process 232, the probabilistic sampler 230 estimates one or more parameters using the posterior probability distribution function. The probabilistic sampler 230 operates on the state and model parameter probability distribution functions from the state and model parameter updaters 224, 226, respectively or alternatively operates on the joint parameter probability distribution function and calculates an output. The output is optionally:

-   -   the state or joint parameter PDF, passed to the PDF resampler         520; and/or;     -   output values resulting from an operation on the inputs to the         output 150 or output display or to the 110 controller.

In one example, expectation values such as a mean and a standard deviation of a state parameter are calculated from the state parameter PDF and output to the user, such as for diagnosis. In another example, expectation values, such as a mean value of state and model parameters, are calculated and then used in a model to output a more advanced diagnostic or prognostic parameter. In a third example, expectation values are calculated on a PDF that is the result of an operation on the state parameter PDF and/or model parameter PDF. Optionally, the output is to the same parameter as the state parameter PDF or model parameter PDF. Other data, such as user-input data, is optionally used in the output operation. The estimated parameters of the probabilistic sampler 230 are optionally used as a feedback to the dynamic state-space model 210 or are used to estimate a biomedical parameter. The feedback to the dynamic state-space model 210 is also referred to as a new probability distribution function or as a new PDF, which is/are updates of the initial state parameter 312 and/or are updates of the initial model parameter 314. Again, for clarity, an example of an estimated parameter 232 is a measurement of the heart/cardiovascular system, such as a heartbeat stroke volume.

Dual Estimator

In another embodiment, the probabilistic updater 220 of the probabilistic digital signal processor 200 uses a dual or joint estimator 222. Referring now to FIG. 4, the joint estimator 222 or dual estimation process uses both a state parameter updater 224 and a model parameter updater 226. Herein, for clarity, a dual estimator 222 is described. However, the techniques and steps described herein for the dual estimator are additionally applicable to a joint estimator as the state parameter and model parameter vector and/or matrix of the dual estimator are merely concatenated in a joint parameter vector and/or are joined in a matrix in a joint estimator.

State Parameter Updater

A first computational model used in the probabilistic updater 220 includes one or more state variables or state parameters, which correspond to the parameter being estimated by the state parameter updater 224. In the case of the hemodynamics monitoring apparatus, state parameters include time, intensity, reflectance, and/or a pressure. Some or all state parameters are optionally selected such that they represent the “true” value of noisy timed sensor data. In this case, calculation of such a posterior state parameter PDF constitutes a noise filtering process and expectation values of the PDF optionally represent filtered sensor values and associated confidence intervals.

Model Parameter Updater

A second computational model used in the probabilistic updater 220 includes one or more model parameters updated in the model parameter updater 226. For example, in the case of the hemodynamics monitoring apparatus, model parameters include: a time interval, a heart rate, a stroke volume, and/or a blood oxygenation percentage.

Hence, the dual estimator 222 optionally simultaneously or in a processing loop updates or calculates one or both of the state parameters and model parameters. The probabilistic sampler 230 is used to determine the estimated value for the biomedical parameter, which is optionally calculated from a state parameter, a model parameter, or a combination of one or more of the state parameter and/or the model parameter.

Referring still to FIGS. 3 and 4 and now referring to FIG. 5, a first example of the dual estimator 222 is described and placed into context of the dynamic state-space model 210 and probabilistic sampler 230 of the probabilistic digital signal processor 200. The state parameter updater 224 element of the dual estimator 222 optionally:

-   -   uses a sensor data integrator 320 operating on the prior PDF         being passed from the dynamic state-space model 210 and         optionally operates on new timed sensor data 122, to produce the         posterior PDF passed to the probabilistic sampler 230;     -   operates on current model parameters 510; and/or     -   in a process 520, the state parameter updater 224 optionally         re-samples a probability distribution function passed from the         probabilistic sampler 230 to form the new probability         distribution function passed to the dynamic state-space model         210.

In addition, in a process 530 the model parameter updater 226 optionally integrates new timed sensor data 122 with output from the probabilistic sampler 230 to form new input to the dynamic state-space model 210.

Referring now to FIG. 6, a second example of a dual estimator 222 is described. In this example:

-   -   initial state parameter probability distribution functions 312         are passed to the dynamic state-space model 210; and/or     -   initial model parameter probability distribution functions 314         are passed to the dynamic state-space model 210.

Further, in this example:

-   -   a Bayesian rule applicator 322 is used as an algorithm in the         sensor data integrator 320;     -   a posterior distribution sample algorithm 522 is used as the         algorithm in the resampling of the PDF process 520; and     -   a supervised or unsupervised machine learning algorithm 532 is         used as the algorithm in the model parameter updater 530.         Filtering

In various embodiments, algorithms, data handling steps, and/or numerical recipes are used in a number of the steps and/or processes herein. The inventor has determined that several algorithms are particularly useful: sigma point Kalman filtering, sequential Monte Carlo filtering, and/or use of a sampler. In a first example, either the sigma point Kalman filtering or sequential Monte Carlo algorithms are used in generating the probability distribution function. In a second example, either the sigma point Kalman filtering or sequential Monte Carlo algorithms are used in the unsupervised machine learning 532 step in the model parameter updater 530 to form an updated model parameter. The sigma point Kalman filtering, sequential Monte Carlo algorithms, and use of a sampler are further described, infra.

Sigma Point Kalman Filter

Filtering techniques based on Kalman and extended Kalman techniques offer advantages over conventional methods and work well for filtering linear systems or systems with small nonlinearities and Gaussian noise. These Kalman filters, however, are not optimum for filtering highly nonlinear systems and/or non-Gaussian/non-stationary noise. In stark contrast, sigma point Kalman filters are well suited to data having nonlinearities and non-Gaussian noise.

Herein, a sigma point Kalman filter (SPKF) refers to a filter using a set of weighted sigma-points that are deterministically calculated, such as by using the mean and square-root decomposition, or an equivalent, of the covariance matrix of a probability distribution function to about capture or completely capture at least the first and second order moments. The sigma-points are subsequently propagated in time through the dynamic state-space model 210 to generate a prior sigma-point set. Then, prior statistics are calculated using tractable functions of the propagated sigma-points, weights, and new measurements.

Sigma point Kalman filter advantages and disadvantages are described herein. A sigma point Kalman filter interprets a noisy measurement in the context of a mathematical model describing the system and measurement dynamics. This gives the sigma point Kalman filter inherent superior performance to all “model-less” methods, such as Wiener filtering, wavelet de-noising, principal component analysis, independent component analysis, nonlinear projective filtering, clustering methods, adaptive noise cancelling, and many others.

A sigma point Kalman filter is superior to the basic Kalman filter, extended Kalman filter, and related variants of the Kalman filters. The extended Kalman filter propagates the random variable using a single measure, usually the mean, and a first order Taylor expansion of the nonlinear dynamic state-space model 210. Conversely, a sigma point Kalman filter decomposes the random variable into distribution moments and propagates those using the unmodified nonlinear dynamic state-space model 210. As a result, the sigma point Kalman filter yields higher accuracy with equal algorithm complexity, while also being easier to implement in practice.

In the sigma-point formalism the probability distribution function is represented by a set of values called sigma points, those values represent the mean and other moments of the distribution which, when input into a given function, recovers the probability distribution function.

Sequential Monte Carlo

Sequential Monte Carlo (SMC) methods approximate the prior probability distribution function through use of a set of weighted sample values without making assumptions about its form. The samples are then propagated in time through the unmodified dynamic state-space model 210. The resulting samples are used to update the posterior via Bayes rule and the latest noisy measurement or timed sensor data 122.

In the sequential Monte Carlo formalism the PDF is actually discretized into a collection of probability “particles” each representing a segment of the probability density in the probability distribution function.

SPKF and SMC

In general, sequential Monte Carlo methods have analysis advantages compared to the sigma point Kalman filters, but are more computationally expensive. However, the SPKF uses a sigma-point set, which is an exact representation only for Gaussian probability distribution functions (PDFs). As a result, SPKFs lose accuracy when PDFs depart heavily from the Gaussian form, such as with bimodal, heavily-tailed, or nonstationary distributions. Hence, both the SMC and SPKF filters have advantages. However, either a SMC analysis or SPKF is used to propagate the prior using the unmodified DSSM. Herein, generally when a SMC filter is used a SPKF filter is optionally used and vise-versa.

A SPKF or a SMC algorithm is used to generate a reference signal in the form of a first probability distribution from the model's current (time=t) physiological state. The reference signal probability distribution and a probability distribution generated from a measured signal from a sensor at a subsequent time (time=t+n) are convoluted using Bayesian statistics to estimate the true value of the measured physiological parameter at time=t+n. The probability distribution function is optionally discrete or continuous. The probability distribution function is optionally used to identify the probability of each value of an unidentified random variable, such as in a discrete function, or the probability of the value falling within a particular interval, such as in a continuous function.

Sampler

Probability distribution functions (PDFs) are optionally continuous or discrete. In the continuous case the probability distribution function is represented by a function. In the discrete case, the variable space is binned into a series of discrete values. In both the continuous and discrete cases, probability distribution functions are generated by first decomposing the PDF into a set of samplers that are characteristic of the probability distribution function and then the samplers are propagated via computations through the DSSM (prior generation) and sensor data integrator (posterior generation). Herein, a sampler is a combination of a value and label. The value is associated with the x-axis of the probability distribution function, which denotes state, model, or joint parameters. The label is associated with the y-axis of the probability distribution function, which denotes the probability. Examples of labels are: weight, frequency, or any arbitrary moment of a given distribution, such as a first Gaussian moment. A powerful example of characteristic sampler use is decomposing the PDF into a series of state values with attached first Gaussian moment labels. This sum of several Gaussian distributions with different values and moments usually gives accurate approximations of the true probability distribution function.

Probabilistic Digital Signal Processor

As described, supra, in various embodiments, the probabilistic digital signal processor 200 comprises one or more of a dynamic state-space model 210, a dual or joint estimator 222, and/or a probabilistic sampler 230, which processes input data, such as sensor data 122 and generates an output 150. Preferably, the probabilistic digital signal processor 200 (1) iteratively processes the data and/or (2) uses a physical model in processing the input data.

The probabilistic digital signal processor 200 optionally:

-   -   filters input data;     -   operates using data from a medical meter, where the medical         meter yields a first physical parameter from raw data, to         generate a second physical parameter not output by the medical         meter;     -   operates on discrete/non-probabilistic input data from a medical         device to generate a probabilistic output function;     -   iteratively circulates a probability distribution function         through at least two of the dynamic state-space model, the dual         or joint updater, and/or the probabilistic sampler;     -   fuses or combines output from multiple medical devices; and/or     -   prognosticates probability of future events.

A hemodynamics example of a probabilistic digital signal processor 200 operating on data from a pulse oximeter is used to describe these processes, infra.

Dynamic State-Space Model

The dynamic state-space model 210 is further described herein.

Referring now to FIG. 7, schematics of an exemplary dynamic state-space model 210 (DSSM) used in the processing of data is provided. The dynamic state-space model 210 typically and optionally includes a process model 710 and/or an observation model 720. The process model 710, F, which mathematically represents mechanical processes involved in generating one or more biomedical parameters, is measured by a sensor, such as as a sensor sensing a mechanical component and describes the state of the biomedical apparatus, output of the biomedical apparatus, and/or state of the patient over time in terms of state parameters. This mathematical model optimally includes mathematical representations accounting for process noise 750, such as mechanically caused artifacts that may cause the sensor to produce a digital output that does not produce an accurate measurement for the biomedical parameter being sensed. The dynamic state-space model 210 also comprises an observational model 720, H, which mathematically represents processes involved in collecting sensor data measured by the mechanical sensor. This mathematical model optimally includes mathematical representations accounting for observation noise produced by the sensor apparatus that may cause the sensor to produce a digital output that does not produce an accurate measurement for a biomedical parameter being sensed. Noise terms in the mathematical models are not required to be additive.

While the process and observation mathematical models 710, 720 are optionally conceptualized as separate models, they are preferably integrated into a single mathematical model that describes processes that produce a biomedical parameter and processes involved in sensing the biomedical parameter. The integrated process and observation model, in turn, is integrated with a processing engine within an executable program stored in a data processor, which is configured to receive digital data from one or more sensors and to output data to a display and/or to another output format.

Still referring to FIG. 7, inputs into the dynamic state-space model 210 include one or more of:

-   -   state parameters 730, such as the initial state parameter         probability distribution function 312 or the new PDF;     -   model parameters 740, such as the initial noise parameter         probability distribution function 314 or an updated model         parameter from the unsupervised machine learning module 532;     -   process noise 750; and/or     -   observation noise 760.         Hemodynamics Dynamic State-Space Model

A first non-limiting specific example is used to facilitate understanding of the dynamic state-space model 210. Referring now to FIG. 8, a hemodynamics dynamic state-space model 805 flow diagram is presented. Generally, the hemodynamics dynamic state-space model 805 is an example of a dynamic state-space model 210. The hemodynamics dynamic state-space model 805 combines sensor data 122, such as a spectral readings of skin, with a physical parameter based probabilistic model. The hemodynamics dynamic state-space model 805 operates in conjunction with the probabilistic updater 220 to form an estimate of heart/cardiovascular state parameters.

To facilitate description of the probabilistic digital signal processor, a non-limiting example of a hemodynamics process model is provided. In this example, the probabilistic digital signal processor is provided:

-   -   raw sensor data, such as current, voltage, and/or resistance;         and/or     -   a first physical parameter output from a medical device.

In this example, the medical device is a pulse oximeter collecting raw data and the first physical parameter from the pulse oximeter provided as input to the probabilistic digital signal processor is one or more of:

-   -   a heart rate; and/or     -   a blood oxygen saturation.

The probabilistic digital signal processor uses a physical model, such as a probabilistic model, to operate on the first physical parameter and/or the raw data to generate a second physical parameter, where the second physical parameter is optionally not the first physical parameter. For example, the output of the probabilistic digital signal processor using a physical hemodynamic model, when provided with the pulse oximeter data, is one or more of:

-   -   a heart stroke volume;     -   a cardiac output flow rate;     -   an aortic blood pressure; and/or     -   a radial blood pressure.

Optionally, the output from the probabilistic model is an updated, error filtered, and/or smoothed version of the original input data, such as a smoothed blood oxygen saturation percentage as a function of time.

Still referring to FIG. 8, to facilitate description of the hemodynamics dynamic state-space model 805, a non-limiting example is provided. In this example, the hemodynamics dynamic state-space model 805 is further described. The hemodynamics dynamic state-space model 805 preferably includes a hemodynamics process model 810 corresponding to the dynamic state-space model 210 process model 710. Further, the hemodynamics dynamic state-space model 805 preferably includes a hemodynamics observation model 820 corresponding to the dynamic state-space model 210 observation model 720. The hemodynamics process model 810 and hemodynamics observation model 820 are further described, infra.

Still referring to FIG. 8, the hemodynamics process model 810 optionally includes one or more of a heart model 812, a vascular model 814, and/or a light scattering or light absorbance model 816. The heart model 812 is a physics based probabilistic model of the heart and movement of blood in and/or from the heart. The vascular model 814 is a physics based probabilistic model of movement of blood in arteries, veins, and/or capillaries. The various models optionally share information. For example, blood flow or stroke volume exiting the heart in the heart model 812 is optionally an input to the arterial blood in the vascular model 814. The light scattering and/or absorbance model 816 relates spectral information, such as from a pulse oximeter, to additional hemodynamics dynamic state-space model parameters, such as heart rate (HR), stroke volume (SV), and/or whole-blood oxygen saturation (SpO₂) or oxyhemoglobin percentage.

Still referring to FIG. 8, the hemodynamics observation model 820 optionally includes one or more of a sensor dynamics and noise model 822 and/or a spectrometer signal transduction noise model 824. Each of the sensor dynamics and noise model 822 and the spectrometer signal transduction noise model 824 are physics based probabilistic models related to noises associated with the instrumentation used to collect data, environmental influences on the collected data, and/or noise due to the human interaction with the instrumentation, such as movement of the sensor. As with the hemodynamics process model 810, the sub-models of the hemodynamics observation model 820 optionally share information. For instance, movement of the sensor noise is added to environmental noise. Optionally and preferably, the hemodynamics observation model 820 shares information with and/provides information to the hemodynamics process model 810.

The hemodynamics dynamic state-space model 805 receives inputs, such as one or more of:

-   -   hemodynamics state parameters 830;     -   hemodynamics model parameters 840;     -   hemodynamics process noise 850; and     -   hemodynamics observation noise 860.

Examples of hemodynamics state parameters 830, corresponding to state parameters 730, include: radial pressure (P_(w)), aortic pressure (P_(ao)), time (t), a spectral intensity (I) or a related absorbance value, a reflectance or reflectance ratio, such as a red reflectance (R_(r)) or an infrared reflectance (R_(ir)), and/or a spectral intensity ratio (I_(R)). Examples of hemodynamics model parameters 840, corresponding to the more generic model parameters 740, include: heart rate (HR), stroke volume (SV), and/or whole-blood oxygen saturation (SpO₂). In this example, the output of the hemodynamics dynamic state-space model 805 is a prior probability distribution function with parameters of one or more of the input hemodynamics state parameters 830 after operation on by the heart dynamics model 812, a static number, and/or a parameter not directly measured or output by the sensor data. For instance, an input data stream is optionally a pulse oximeter yielding spectral intensities, ratios of intensities, and a percent oxygen saturation. However, the output of the hemodynamics dynamic state-space model is optionally a second physiological value, such as a stroke volume of the heart, which is not measured by the input biomedical device.

The hemodynamics dynamic state-space model 805 optionally receives inputs from one or more additional models, such as an irregular sampling model, which relates information collected at irregular or non-periodic intervals to the hemodynamics dynamic state-space model 805.

Generally, the hemodynamics dynamic state-space model 805 is an example of a dynamic state-space model 210, which operates in conjunction with the probabilistic updater 220 to form an estimate of a heart state parameter and/or a cardiovascular state parameter.

Generally, the output of the probabilistic signal processor 200 optionally includes a measure of uncertainty, such as a confidence interval, a standard deviation, and/or a standard error. Optionally, the output of the probabilistic signal processor 200 includes:

-   -   a filtered or smoothed version of the parameter measured by the         medical meter; and/or     -   a probability function associated with a parameter not directly         measured by the medical meter.

Example I

An example of a pulse oximeter with probabilistic data processing is provided as an example of the hemodynamics dynamic state-space model 805. The model is suitable for processing data from a pulse oximeter model. In this example, particular equations are used to further describe the hemodynamics dynamic state-space model 805, but the equations are illustrative and non-limiting in nature.

Heart Model

An example of the heart model 812 is used to further described an example of the hemodynamics dynamic state-space model 805. In this example, cardiac output is represented by equation 1,

$\begin{matrix} {{Q_{CO}(t)} = {{\overset{\_}{Q}}_{CO}{\sum\limits_{1}^{\delta}{a_{k}{\exp\left\lbrack \frac{- \left( {t - b_{k}} \right)^{2}}{c_{k}^{2}} \right\rbrack}}}}} & (1) \end{matrix}$ where cardiac output Q_(co)(t), is expressed as a function of heart rate (HR) and stroke volume (SV) and where Q_(co)=(HR×SV)/60. The values a_(k), b_(k), and c_(k) are adjusted to fit data on human cardiac output. Vascular Model

An example of the vascular model 814 of the hemodynamics state-space model 805 is provided. The cardiac output function pumps blood into a Windkessel 3-element model of the vascular system including two state variables: aortic pressure, P_(ao), and radial (Windkessel) pressure, P_(w), according to equations 2 and 3,

$\begin{matrix} {P_{w,{k + 1}} = {{\frac{1}{C_{w}R_{p}}\left( {{\left( {R_{P} + Z_{0}} \right)Q_{CO}} - P_{{CO},k}} \right)\delta\; t} + P_{w,k}}} & (2) \\ {P_{{ao},{k + 1}} = {P_{w,{k + 1}} + {Z_{0}Q_{CO}}}} & (3) \end{matrix}$ where R_(p) and Z₀ are the peripheral resistance and characteristic aortic impedance, respectively. The sum of these two terms is the total peripheral resistance due to viscous (Poiseuille-like) dissipation according to equation 4, Z ₀=√{square root over (ρ/AC ₁)}  (4) where ρ is blood density and C_(l) is the compliance per unit length of artery. The elastic component due to vessel compliance is a nonlinear function including thoracic aortic cross-sectional area, A: according to equation 5,

$\begin{matrix} {{A\left( P_{CO} \right)} = {A_{m\;{ax}}\left\lbrack {\frac{1}{2} + {\frac{1}{\pi}{\arctan\left( \frac{P_{CO} - P_{0}}{P_{1}} \right)}}} \right\rbrack}} & (5) \end{matrix}$ where A_(max), P₀, and P₁ are fitting constants correlated with age and gender according to equations 6-8. A _(max)=(5.62−1.5(gender))·cm²  (6) P ₀=64(gender)−0.89(age))·mmHg  (7) P ₁(57−0.44(age))·mmHg  (8)

The time-varying Windkessel compliance, C_(w), and the aortic compliance per unit length, C_(l), are related in equation 9,

$\begin{matrix} {C_{w} = {{IC}_{l} = {{l\;\frac{dA}{{dP}_{\infty}}} = {l\;\frac{A_{{ma}\; x}/\left( {\pi\; P_{1}} \right)}{1 + \left( \frac{P_{\infty} - P_{0}}{P_{1}} \right)}}}}} & (9) \end{matrix}$ where l is the aortic effective length. The peripheral resistance is defined as the ratio of average pressure to average flow. A set-point pressure, P_(set), and the instantaneous flow related to the peripheral resistance, R_(p), according to equation 10,

$\begin{matrix} {R_{P} = \frac{P_{set}}{\left( {{HR} \cdot {SV}} \right)/60}} & (10) \end{matrix}$ are used to provide compensation to autonomic nervous system responses. The value for P_(set) is optionally adjusted manually to obtain 120 over 75 mmHg for a healthy individual at rest. Light Scattering and Absorbance Model

The light scattering and absorbance model 816 of the hemodynamics dynamic state-space model 805 is further described. The compliance of blood vessels changes the interactions between light and tissues with pulse. This is accounted for using a homogenous photon diffusion theory for a reflectance or transmittance pulse oximeter configuration according to equation 11,

$\begin{matrix} {R = {\frac{I_{a\; c}}{I_{d\; c}} = {\frac{\Delta\; I}{I} = {\frac{3}{2}{\sum\limits_{s}^{1}{{K\left( {\alpha,d,r} \right)}{\sum\limits_{a}^{art}{\Delta\; V_{0}}}}}}}}} & (11) \end{matrix}$ for each wavelength. In this example, the red and infrared bands are centered at about 660±100 nm and at about 880±100 nm. In equation 11, I (no subscript) denotes the detected intensity, R, is the reflected light, and the alternating current intensity, I_(ac), is the pulsating signal, ac intensity, or signal; and the background intensity, I_(dc), is the direct current intensity or dc intensity; α, is the attenuation coefficient; d, is the illumination length scale or depth of photon penetration into the skin; and r is the distance between the source and detector.

Referring again to the vascular model 814, V_(a) is the arterial blood volume, which changes as the cross-sectional area of illuminated blood vessels, ΔA_(w), according to equation 12, ΔV _(a) ≈r·ΔA _(w)  (12) where r is the source-detector distance.

Referring again to the light scattering and absorbance model 816, the tissue scattering coefficient, Σ′_(s), is assumed constant but the arterial absorption coefficient, Σ_(a) ^(art), which represents the extinction coefficients, depends on blood oxygen saturation, SpO₂, according to equation 13,

$\begin{matrix} {\overset{art}{\sum\limits_{a}}{= {\frac{H}{v_{i}}\left\lbrack {{{SpO}_{2} \cdot \sigma_{0}^{100\%}} + {\left( {1 - {SpO}_{2}} \right) \cdot \sigma_{0}^{0\%}}} \right\rbrack}}} & (13) \end{matrix}$ which is the Beer-Lambert absorption coefficient, with hematocrit, H, and red blood cell volume, v_(i). The optical absorption cross-sections, proportional to the absorption coefficients, for red blood cells containing totally oxygenated (HbO₂) and totally deoxygenated (Hb) hemoglobin are σ_(a) ^(100%) and σ_(a) ^(0%), respectively.

The function K(α, d, r), along with the scattering coefficient, the wavelength, sensor geometry, and oxygen saturation dependencies, alters the effective optical pathlengths, according to equation 14.

$\begin{matrix} {{K\left( {\alpha,d,r} \right)} \approx \frac{- r^{2}}{1 + {\alpha\; r}}} & (14) \end{matrix}$

The attenuation coefficient α is provided by equation 15, α=√{square root over (3Σ_(a)(Σ_(s)+Σ_(a)))}  (15) where Σ_(a) and Σ_(s) are whole-tissue absorption and scattering coefficients, respectively, which are calculated from Mie Theory.

Red, K_(r) , and infrared, K_(ir) , K values as a function of SpO₂ are optionally represented by two linear fits, provided in equations 16 and 17 K _(r) ≈−4.03·SpO₂−1.17  (16) K _(ir) ≈0.102·SpO₂−0.753  (17) in mm². The overbar denotes the linear fit of the original function. Referring yet again to the vascular model 814, the pulsatile behavior of ΔA_(w), which couples optical detection with the cardiovascular system model, is provided by equation 18,

$\begin{matrix} {{\Delta\; A_{w}} = {\frac{A_{w,{{ma}\; x}}}{\pi}\frac{P_{w,1}}{P_{w,1}^{2} + \left( {P_{w,{k + 1}} - P_{w,0}} \right)^{2}}\Delta\; P_{w}}} & (18) \end{matrix}$ where P_(w,0)=(⅓)P₀ and P_(w,1)=(⅓)P₁ account for the poorer compliance of arterioles and capillaries relative to the thoracic aorta. The subscript k is a data index and the subscript k+1 or k+n refers to the next or future data point, respectively.

Referring yet again to the light scattering and absorbance models, third and fourth state variables, the red and infrared reflected intensity ratios, R=I_(ac)/I_(dc), are provided by equations 19 and 20. R _(r,k+1) =cΣ′ _(s,y) K _(r) Σ_(a,y) ^(art) ΔA _(w) +R _(r,k) +v _(r)  (19) R _(ir,k+1) =cΣ′ _(s,ir) K _(ir) Σ_(a,ir) ^(art) +ΔA _(w) +R _(ir,k) +v _(ir)  (20) Here, v is a process noise, such as an added random number or are Gaussian-distributed process noises intended to capture the baseline wander of the two channels, Σ′_(s,r) and Σ′_(s,ir) are scattering coefficients, and Σ′_(a,r) ^(art) and Σ′_(a,ir) ^(art) are absorption coefficients. Sensor Dynamics and Noise Model

The sensor dynamics and noise model 822 is further described. The constant c subsumes all factors common to both wavelengths and is treated as a calibration constant. The observation model adds noises, n, with any probability distribution function to R_(r) and R_(ir), according to equation 21.

$\begin{matrix} {\begin{bmatrix} y_{r,k} \\ y_{{ir},k} \end{bmatrix} = {\begin{bmatrix} R_{r,k} \\ R_{{ir},k} \end{bmatrix} + \begin{bmatrix} n_{r,k} \\ n_{{ir},k} \end{bmatrix}}} & (21) \end{matrix}$

A calibration constant, c, was used to match the variance of the real I_(ac)/I_(dc) signal with the variance of the dynamic state-space model generated signal for each wavelength. After calibration, the age and gender of the patient was entered. Estimates for the means and covariances of both state and parameter PDFs are optionally entered.

Referring now to FIG. 9, processed data from a relatively high signal-to-noise ratio pulse oximeter data source is provided for about a fifteen second stretch of data. Referring now to FIG. 9A, input photoplethysmographic waveforms are provided. Using the hemodynamics dynamic state-space model 805, the input waveforms were used to extract heart rate (FIG. 9B), left-ventricular stroke volume (FIG. 9C), cardiac output (FIG. 9D), blood oxygen saturation (FIG. 9E), and aortic and systemic (radial) pressure waveforms (FIG. 9F). Several notable points are provided. First, the pulse oximeter provided a first physical value of a hemoglobin oxygen saturation percentage. However, the output blood oxygen saturation percentage, FIG. 9E, was processed by the probabilistic digital signal processor 200. Due to the use of the sensor dynamics and noise model 822 and the spectrometer signal transduction noise model, noisy data, such as due to ambulatory movement of the patient, is removed in the smoothed and filtered output blood oxygen saturation percentage. Second, some pulse oximeters provide a heart rate. However, in this case the heart rate output was calculated using the physical probabilistic digital signal processor 200 in the absence of a heart rate input data source 122. Third, each of the stroke volume, FIG. 9C, cardiac output flow rate, FIG. 9D, aortic blood pressure, FIG. 9E, and radial blood pressure, FIG. 9E, are second physical parameters that are different from the first physical parameter measured by the pulse oximeter photoplethysmographic waveforms.

Referring now to FIG. 10, a second stretch of photoplethysmographic waveforms are provided that represent a low signal-to-noise ratio signal from a pulse oximeter. Low signal-to-noise photoplethysmographic waveforms (FIG. 10A) were used to extract heart rate (FIG. 10B), left-ventricular stroke volume (FIG. 10C), blood oxygen saturation (FIG. 10D), and aortic and systemic (radial) pressure waveforms (FIG. 10E) using the hemodynamics dynamic state-space model 805. In each case, the use of the probabilistic digital signal processor 200 configured with the optional sensor dynamics and noise model 822 and spectrometer signal transduction model 824 overcame the noisy input stream to yield smooth and functional output data for medical use.

The various models relate measurement parameters from a source medical device to a second parameter not measured by the source medical device. For example, an oxygen level is related to a heart stroke volume.

Electrocardiography

Electrocardiography is a noninvasive transthoracic interpretation of the electrical activity of the heart over time as measured by externally positioned skin electrodes. An electrocardiographic device produces an electrocardiogram (ECG or EKG).

The electrocardiographic device operates by detecting and amplifying the electrical changes on the skin that are caused when the heart muscle depolarizes, such as during each heartbeat. At rest, each heart muscle cell has a charge across its outer wall or cell membrane. Reducing the charge toward zero is called de-polarization, which activates the mechanisms in the cell that cause it to contract. During each heartbeat a healthy heart will has orderly progression of a wave of depolarization that is triggered by the cells in the sinoatrial node, spreads out through the atrium, passes through intrinsic conduction pathways, and then spreads all over the ventricles. The conduction is detected as increases and decreases in the voltage between two electrodes placed on either side of the heart. The resulting signal is interpreted in terms of heart health, function, and/or weakness in defined locations of the heart muscles.

Examples of electrocardiograph device lead locations and abbreviations include:

-   -   right arm (RA);     -   left arm (LA);     -   right leg (RL);     -   left leg (LL);     -   in fourth intercostal space to right of sternum (V₁);     -   in fourth intercostal space to left of the sternum (V₂);     -   between leads V₂ and V₄ (V₃);     -   in the fifth intercostal space in the mid clavicular line (V₄);     -   horizontally even with V₄, but in the anterior axillary line         (V₅); and     -   horizontally even with V₄ and V₅ in the midaxillary line (V₆).

Usually more than two electrodes are used and they are optionally combined into a number of pairs. For example, electrodes placed at the left arm, right arm, and left leg form the pairs LA+RA, LA+LL, and RA+LL. The output from each pair is known as a lead. Each lead examines the heart from a different angle. Different types of ECGs can be referred to by the number of leads that are recorded, for example 3-lead, 5-lead, or 12-lead ECGs.

Electrocardiograms are used to measure and diagnose abnormal rhythms of the heart, such as abnormal rhythms caused by damage to the conductive tissue that carries electrical signals or abnormal rhythms caused by electrolyte imbalances. In a myocardial infarction (MI) or heart attack, the electrocardiogram is used to identify if the heart muscle has been damaged in specific areas. Notably, traditionally an ECG cannot reliably measure the pumping ability of the heart, for which additional tests are used, such as ultrasound-based echocardiography or nuclear medicine tests. Along with other uses of an electrocardiograph model, the probabilistic mathematical electrocardiograph model, described infra, shows how this limitation is overcome.

Example II

A second example of a dynamic state-space model 210 coupled with a dual or joint estimator 222 and/or a probabilistic updater 220 or probabilistic sampler 230 in a medical or biomedical application is provided.

Ischemia and Heart Attack

For clarity, a non-limiting example of prediction of ischemia using an electrocardiograph dynamic state-space model is provided. A normal heart has stationary and homogenous myocardial conducting pathways. Further, a normal heart has stable excitation thresholds resulting in consecutive beats that retrace with good fidelity. In an ischemic heart, conductance bifurcations and irregular thresholds give rise to discontinuous electrophysiological characteristics. These abnormalities have subtle manifestations in the electrocardiograph morphology that persist long before shape of the electrocardiograph deteriorates sufficiently to reach detection by a skilled human operator. Ischemic abnormalities are characterized dynamically by non-stationary variability between heart beats, which are difficult to detect, especially when masked by high frequency noise, or similarly non-stationary artifact noise, such as electrode lead perturbations induced by patient motion.

Detection performance is improved substantially relative to the best practitioners and current state-of-the-art algorithms by integrating a mathematical model of the heart with accurate and rigorous handling of probabilities. An example of an algorithm for real time and near-optimal ECG processing is the combination of a sequential Monte Carlo algorithm with Bayes rule. Generally, an electrodynamic mathematical model of the heart with wave propagation through the body is used to provide a “ground truth” for the measured signal from the electrocardiograph electrode leads. Use of a sequential Monte Carlo algorithm predicts a multiplicity of candidate values for the signal, as well as other health states, at each time point, and each is used as a prior to calculate the truth estimate based on sensor input via a Bayesian update rule. Since the model is electrodynamic and contains state and model parameter variables corresponding to a normal condition and an ischemic condition, such events can be discriminated by the electrocardiograph model, described infra.

Unlike simple filters and algorithms, the electrocardiograph dynamic state-space model coupled with the probabilistic updater 220 or probabilistic sampler 230 is operable without the use of assumptions about the regularity of morphological variation, spectra of noise or artifact, or the linearity of the heart electrodynamic system. Instead, the dynamic response of the normal or ischemic heart arises naturally in the context of the model during the measurement process. The accurate and rigorous handling of probabilities of this algorithm allows the lowest possible detection limit and false positive alarm rate at any level of noise and/or artifact corruption.

Electrocardiograph with Probabilistic Data Processing

FIG. 11 is a schematic of an electrocardiograph dynamic state-space model suitable for processing electrocardiogram data, including components required to describe the processes occurring in a subject. The combination of SPKF or SMC filtering in state, joint, or dual estimation modes is optionally used to filter electrocardiograph (ECG) data. Any physiology model adequately describing the ECG signal is optionally used, as well as any model of noise and artifact sources interfering or contaminating the signal. One non-limiting example of such a model is a model using a sum of arbitrary wave functions with amplitude, center and width, respectively, for each wave (P, Q, R, S, T) in an ECG. The observation model comprises the state plus additive Gaussian noise, but more realistic pink noise or any other noise probability distributions is optionally used.

Still referring to FIG. 11, to facilitate description of the electrocardiograph dynamic state-space model 1105, a non-limiting example is provided. In this example, the electrocardiograph dynamic state-space model 1105 is further described. The electrocardiograph dynamic state-space model 1105 preferably includes a heart electrodynamics model 1110 corresponding to the dynamic state-space model 210 process model 710. Further, the electrocardiograph dynamic state-space model 1105 preferably includes a heart electrodynamics observation model 1120 corresponding to the dynamic state-space model 210 observation model 720. The electrocardiograph process model 1110 and electrocardiogram observation model 1120 are further described, infra.

Still referring to FIG. 11, the electrocardiograph process model 1110 optionally includes one or more of a heart electrodynamics model 1112 and a wave propagation model 1114. The heart electrodynamics model 1112 is a physics based model of the electrical output of the heart. The wave propagation model 1114 is a physics based model of movement of the electrical pulses through the lungs, fat, muscle, and skin. An example of a wave propagation model 1114 is a thorax wave propagation model modeling electrical wave movement in the chest, such as through an organ. The various models optionally share information. For example, the electrical pulse of the heart electrodynamics model 1112 is optionally an input to the wave propagation model 1114, such as related to one or more multi-lead ECG signals. Generally, the process model 710 components are optionally probabilistic, but are preferentially deterministic. Generally, the observation model 720 components are probabilistic.

Still referring to FIG. 11, the electrocardiogram observation model 1120 optionally includes one or more of a sensor noise and interference model 1122, a sensor dynamics model 1124, and/or an electrode placement model 1126. Each of the sensor noise and interference model 1122 and the sensor dynamics models 1124 are optionally physics based probabilistic models related to noises associated with the instrumentation used to collect data, environmental influences on the collected data, and/or noise due to the human interaction with the instrumentation, such as movement of the sensor. A physics based model uses at least one equation relating forces, electric fields, pressures, or light intensity to sensor provided data. The electrode placement model 1126 relates to placement of the electrocardiograph leads on the body, such as on the arm, leg, or chest. As with the electrocardiograph process model 1110, the sub-models of the electrocardiograph observation model 1120 optionally share information. For instance, a first source of noise, such as sensor noise related to movement of the sensor, is added to a second source of noise, such as a signal transduction noise. Optionally and preferably, the electrocardiograph observation model 1120 shares information with and/provides information to the electrocardiograph process model 1110.

The electrocardiograph dynamic state-space model 1105 receives inputs, such as one or more of:

-   -   electrocardiograph state parameters 1130;     -   electrocardiograph model parameters 1140;     -   electrocardiograph process noise 1150; and     -   electrocardiograph observation noise 1160.

Examples of electrocardiograph state parameters 1130, corresponding to state parameters 730, include: atrium signals (AS), ventricle signals (VS) and/or ECG lead data. Examples of electrocardiograph model parameters 1140, corresponding to the more generic model parameters 740, include: permittivity, ε, autonomic nervous system (ANS) tone or visceral nervous system, and a heart rate variability (HRV). Heart rate variability (HRV) is a physiological phenomenon where the time interval between heart beats varies and is measured by the variation in the beat-to-beat interval. Heart rate variability is also referred to as heart period variability, cycle length variability, and RR variability, where R is a point corresponding to the peak of the QRS complex of the electrocardiogram wave and RR is the interval between successive Rs. In this example, the output of the electrocardiograph dynamic state-space model 1105 is a prior probability distribution function with parameters of one or more of the input electrocardiograph state parameters 1130 after operation on by the heart electrodynamics model 1112, a static number, a probability function, and/or a parameter not measured or output by the sensor data.

An example of an electrocardiograph with probabilistic data processing is provided as an example of the electrocardiogram dynamic state-space model 1105. The model is suitable for processing data from an electrocardiograph. In this example, particular equations are used to further describe the electrocardiograph dynamic state-space model 1105, but the equations are illustrative and non-limiting in nature.

Heart Electrodynamics

The heart electrodynamics model 1112 of the ECG dynamic state-space model 1105 is further described. The transmembrane potential wave propagation in the heart is optionally simulated using FitzHugh-Nagumo equations. The heart model 1112 is optionally implemented, for instance, as a coarse-grained three-dimensional heart anatomical model or as a compartmental, zero-dimensional model of the heart. The latter could take the form, for instance, of separate atrium and ventricle compartments.

In a first example of a heart electrodynamics model 1112, a first set of equations for cardiac electrodynamics are provided by equations 22 and 23,

$\begin{matrix} {\overset{.}{u} = {{{div}\left( {D\;{\nabla u}} \right)} + {{{ku}\left( {1 - u} \right)}\left( {u - a} \right)} - {uz}}} & (22) \\ {\overset{.}{z} = {{- \left( {e + \frac{u_{1}z}{u + u_{2}}} \right)}\left( {{{ku}\left( {u - a - 1} \right)} + z} \right)}} & (23) \end{matrix}$ where D is the conductivity, u is a normalized transmembrane potential, and z is a secondary variable for the repolarization. In the compartmental model, u_(i) becomes either the atrium potential, u_(as), or the ventricle potential, u_(vs). The repolarization is controlled by k and e, while the stimulation threshold and the reaction phenomenon is controlled by the value of a. The parameters μ₁ and μ₂ are preferably empirically fitted.

A second example of a heart electrodynamics model is presented, which those skilled in the art will understand is related to the first heart electrodynamics model. The second heart electrodynamics model is expanded to include a restitution property of cardiac tissue, where restitution refers to a return to an original physical condition, such as after elastic deformation of heart tissue. The second heart electrodynamics model is particularly suited to whole heart modeling and is configured for effectiveness in computer simulations or models.

The second heart electrodynamics model includes two equations, equations 24 and 25, describing fast and slow processes and is useful in adequately representing the shape of heart action potential,

$\begin{matrix} {\frac{\partial u}{\partial t} = {{\frac{\partial}{\partial x_{i}}d_{ij}\frac{\partial u}{\partial x_{j}}} - {{{ku}\left( {u - a} \right)}\left( {u - 1} \right)} - {uv}}} & (24) \\ {\frac{\partial v}{\partial t} = {{ɛ\left( {u,v} \right)}\left( {{- v} - {{ku}\left( {u - a - 1} \right)}} \right)}} & (25) \end{matrix}$ where ε(u,v)=ε₀+u₁v/(u+u₂). Herein, the approximate values of k=8, a=0.15, and ε₀=0.002 are used, but the values are optionally set for a particular model. The parameters u₁ and u₂ are set for a given model and d_(ij) is the conductivity tensor accounting for the heart tissue anisotropy.

Further, the second heart electrodynamics model involves dimensionless variables, such as u, v, and t. The actual transmembrane potential, E, and time, t, are obtained using equations 26 and 27 or equivalent formulas. e[mV]=100u−80  (26) t[ms]=12.9t[t.u.]  (27)

In this particular case, the rest potential E_(rest) is about −80 mV and the amplitude of the pulse is about 100 mV. Time is scaled assuming a duration of the action potential, APD, measured at the level of about ninety percent of repolarization, APD₀=330 ms. The nonlinear function for the fast variable u optionally has a cubic shape.

The dependence of ε on u and v allows the tuning of the restitution curve to experimentally determined values using u₁ and u₂. The shape of the restitution curve is approximated by equation 28,

$\begin{matrix} {{APD} = \frac{CL}{\left( {{aCL} + b} \right)}} & (28) \end{matrix}$ where the duration of the action potential, APD, is related to the cycle length, CL. In dimensionless form, equation 28 is rewritten according to equation 29,

$\begin{matrix} {\frac{1}{apd} = {1 + \frac{b}{cl}}} & (29) \end{matrix}$ where apd=APD/APD₀, and APD₀ denotes APD of a free propagating pulse.

Restitution curves with varying values of parameters u₁ and u₂ are used, however, optional values for parameters u₁ and u₂ are about u₁=0.2 and u₂=0.3. One form of a restitution curve is a plot of apd vs. cl, or an equivalent. Since a restitution plot using apd vs. cl is a curved line, a linear equivalent is typically preferred. For example, restitution curve is well fit by a straight line according to equation 30.

$\begin{matrix} {\frac{1}{apd} = {k_{1} + \frac{k_{2}}{cl}}} & (30) \end{matrix}$

Optional values of k₁ and k₂ are about 1.0 and 1.05, respectively, but are preferably fit to real data for a particular model. Generally, the parameter k₂ is the slope of the line and reflects the restitution at larger values of CL.

The use of the electrodynamics equations, the restitutions, and/or the restitution curve is subsequently used to predict or measure arrhythmia. Homogeneous output is normal. Inhomogeneous output indicates a bifurcation or break in the conductivity of the heart tissue, which has an anisotropic profile, and is indicative of an arrhythmia. Hence, the slope or shape of the restitution curve is used to detect arrhythmia.

Wave Propagation

The electric wave model 1114 of the ECG dynamic state-space model 1105 is further described. The propagation of the heart electrical impulse through lung and other tissues before reaching the sensing electrodes is optionally calculated using Gauss' Law,

$\begin{matrix} {{\nabla{\cdot {E(t)}}} = \frac{u_{i}(t)}{ɛ_{0}}} & (31) \end{matrix}$ where μ_(i)(t) is the time-varying charge density given by the heart electrodynamics model and ε₀ is the permittivity of free space, which is optionally scaled to an average tissue permittivity. Sensor Dynamics

The sensor dynamics model 1124 of the ECG dynamic state-space model 1105 is further described. The ECG sensor is an electrode that is usually interfaced by a conducting gel to the skin. When done correctly, there is little impedance from the interface and the wave propagates toward a voltage readout. The overall effect of ancillary electronics on the measurement should be small. The relationship between the wave and readout can be written in general as: V(t)=G(E(t))+N(p)+D(s,c)  (32) where G is the map from the electrical field reaching the electrode and voltage readout. This includes the effect of electronics and electrode response timescales, where N is the sensor noise and interference model and D is the electrode placement model. Sensor Noise and Interference Model

The sensor noise and interference model 1122 of the ECG dynamic state-space model 1105 is further described. The sensor noise enters the DSSM as a stochastic term (Langevin) that is typically additive but with a PDF that is both non-Gaussian and non-stationary. Optionally non-stationarity is modeled from the perturbation, p, representing both external interference and cross-talk. One way to accomplish this is to write: N(E(t),p)=αn _(t) +βpn ₂  (33) where alpha, α, and beta, β, are empirical constants and n₁ and n₂ are stochastic parameters with a given probability distribution function. Electrode Placement Model

The electrode placement model 1126 of the ECG dynamic state-space model 1105 is further described. This model is an anatomical correction term to the readout equation operating on the sagittal and coronal coordinates, s and c, respectively. This model varies significantly based on distance to the heart and anatomical structures between the heart and sensor. For instance, the right arm placement is vastly different than the fourth intercostal.

Optionally, the output from the electrocardiograph probabilistic model is an updated, error filtered, or smoothed version of the original input data. For example, the probabilistic processor uses a physical model where the output of the model processes low signal-to-noise ratio events to yield any of: an arrhythmia detection, arrhythmia monitoring, an early arrhythmia warning, an ischemia warning, and/or a heart attack prediction.

Optionally, the model compares shape of the ECG with a reference look-up table, uses an intelligent system, and/or uses an expert system to estimate, predict, or produce one or more of: an arrhythmia detection, an ischemia warning, and/or a heart attack warning.

Referring now to FIG. 12A and FIG. 12B, the results of processing noisy non-stationary ECG signals are shown. Heart rate oscillations representative of normal respiratory sinus arrhythmia are present in the ECG. The processor accomplishes accurate, simultaneous estimation of the true ECG signal and a heart rate that follows closely the true values. Referring now to FIG. 13A and FIG. 13B, the performance of the processor using a noise and artifact-corrupted signal is shown. A clean ECG signal representing one heart beat was contaminated with additive noise and an artifact in the form of a plateau at R and S peaks (beginning at time=10 sec). Estimates by the processor remain close to the true signal despite the noise and artifact.

Fusion Model

Optionally, inputs from multiple data sources, such as sensors or medical instruments, are fused and used in the probabilistic digital signal processor 200. The fused data often include partially overlapping information, which is shared between models, used in a fused model, and/or is used in a global model to enhance parameter estimation. The overlapping information results in benefits of the fused model, including:

-   -   enhanced accuracy of an estimated parameter;     -   enhanced precision of an estimated parameter;     -   noise artifact reduction in a data stream; and/or     -   an additionally determined metric.

Herein, fusion of data from biomedical sensors is used to illustrate the benefits of sensors fusion in combination with a physical model. However, the concept extends to cover mechanical systems using sensors.

Data Fusion

Referring now to FIG. 14, an overview of a sensor fusion system 1400 in combination with at least one physical model and a probabilistic processor 200 is provided. Generally, data from multiple instruments 1405 is provided to the probabilistic processor 200, such as to the probabilistic updater 220, dual or joint estimator 222, state parameter updater 224, and/or the model parameter updater 226. More particularly, data from a first instrument 1410, second instrument 1420, third instrument 1430, and/or n^(th) instrument 1440 is provided to the probabilistic processor 200, where n is a positive integer, such as at least 2, 3, 4, or 5. One or more of the n instruments 1405 optionally include readings from multiple sensors. As a first example, if the first instrument 1410 is a pulse oximeter, then output from the pulse oximeter as input to the probabilistic processor 200 optionally includes one or more of: raw sensor data, voltage data, processed spectral intensity data, or pulse oximeter generated output data, such as a blood oxygen saturation percentage. As a second example, if the second instrument 1420 is an electrocardiograph device, then output from the electrocardiograph device as input to the probabilistic processor 200 optionally includes one or more of: raw sensor data, current, voltage, resistance, processed electrocardiograph device signal, and/or an outcome, such as an indication of a previous heart attack. In a third example, output from an instrument includes environmental information, such as temperature, pressure, vibration, and humidity. Herein, time readings are optionally input along with any of the sensor data from any of the multiple instruments 1405, but time is not considered a sensed value nor does time count as one of the multiple data sources fused with the probabilistic processor 200. The fused sensor data 1450 refers to any form, matrix, concatenation, combination, union, representation, or mathematical combination of the data from the multiple instruments 1405. The fused sensor data 1450 is preferably fused by use of the probabilistic processor 200 but is optionally fused prior to input into the probabilistic processor 200.

Referring now to FIG. 15, an example of a pulse oximeter 1510 and an electrocardiograph meter or device 1520 used as inputs to the probabilistic processor 200 is provided. The pulse oximeter 1510 provides time dependent values to the probabilistic processor 200, such as raw sensor data, voltage data, processed spectral intensity data, or pulse oximeter generated output data, such as a blood oxygen saturation percentage. The electrocardiograph meter 1520 additionally provides time dependent values to the probabilistic processor 200, such as raw sensor data, current, voltage, resistance, processed electrocardiograph device signal, and/or an outcome, such as a previous heart attack indication. The pulse oximeter 1510 data and electrocardiograph device 1520 data are optionally fused, as described supra. As discussed, infra, additional input data is provided to the probabilistic processor 200, such as data from an accelerometer 1530, data from a time meter 1550, and/or data from an environment meter 1540, such as temperature, pressure, vibration, humidity, and/or position information. The data is at least partially fused into fused sensor data 1450, as described supra.

Integration of Fused Data with Probabilistic Processor

Referring now to FIG. 16, an example of data originating from the multiple instruments 1405 as input to the dual or joint estimator 222 is provided. As illustrated, the data from the multiple instruments 1405, described supra, is optionally input into the state parameter updater 224 or into the model parameter updater 226. As described, supra, the data from the multiple instruments 1405 is optionally fused prior to and/or after entry into any of the probabilistic processor 200 sub-components or software algorithms. Similarly, the initial probability distribution function parameters 310 optionally include initial values/probabilities for each of the multiple instruments 1405.

Fusion Configured Dynamic State-Space Model

Referring now to FIG. 17, an example of a dynamic state-space model 210 configured for use with data from the multiple instruments 1405 is provided.

Process Model

For example, the process model 710 of the dynamic state-space model 210, optionally includes a first process model 712 related to data from the first instrument 1410 and a second process model 714 configured to use and represent data from the second instrument 1420. Generally, there are about n process models 716 related to the n instruments 1440, though 1, 2, 3, or more process models are optionally configured to represent or process the data from the n instruments.

Observation Model

Similarly, the observation model 720 of the dynamic state-space model 210, optionally includes a first observation model 722 related to data from the first instrument 1410 and a second observation model 724 configured to use and represent data from the second instrument 1420. Generally, there are about n observation models 716 related to the n instruments 1440, though 1, 2, 3, or more observation models are optionally configured to represent or process the data from the n instruments.

State and Model Parameters

The dynamic state-space model optionally receives state parameter 730 inputs. Examples of DSSM inputs include:

-   -   a first state parameter 732, such as a parameter from the first         instrument 1410;     -   a second state parameter 734, such as a value measured by the         second instrument 1420; and     -   an n^(th) state parameter 736, such as a parameter determined by         the dynamic state-space model 210.

Similarly, the dynamic state-space model 210 optionally receives model parameter 740 inputs. Examples of model parameter inputs include:

-   -   a first model parameter 742, such as a parameter from the first         instrument 1410;     -   a second model parameter 744, such as a modeled value; and     -   an n^(th) state parameter 746, such as a parameter determined by         the dynamic state-space model 210.

The dynamic state-space model 210 optionally receives fusion process noise 750 input and/or fusion observation noise 760 input.

Pulse Oximeter/Electrocardiograph Fusion

The non-limiting example of fusion of information from a pulse oximeter and an electrocardiogram device is further described to clarify model fusion and/or information combination.

A pulse oximeter and an electrocardiograph meter both provide information on the heart. Hence, the pulse oximeter and the electrocardiograph meter provide overlapping information, which is optionally shared, such as between the hemodynamics dynamic state-space model 805 and the electrocardiogram dynamic state-space model 1105. Similarly, a fused model incorporating aspects of both the hemodynamics dynamic state-space model 805 and the electrocardiogram dynamic state-space model 1105 is created, which is an example of a fused model. Particularly, in an electrocardiogram the left-ventricular stroke volume is related to the power spent during systolic contraction, which is, in turn, related to the electrical impulse delivered to that region of the heart. Indeed, the R-wave amplitude is optionally correlated to contractility. It is readily seen that other features of the electrocardiogram also have relationships with the cardiac output function. As described, supra, the pulse oximeter also provides information on contractility, such as heart rate, stroke volume, cardiac output flow rate, and/or blood oxygen saturation information. Since information in common is present, the system is over determined, which allows outlier analysis and/or calculation of a heart state or parameter with increased accuracy and/or precision.

Example I

Referring now to FIG. 18, a particular example of a fused dynamic state-space model 1805 is presented. In this example, output from a traditional pulse oximeter 1510 is fused with output from a traditional electrocardiogram device 1520. In this example, the fused dynamic state-space model 1805 incorporates models covering both hemodynamics and heart electrodynamics. Generally, a fused dynamic state-space model 1805 incorporates one or more models modeling information from the multiple instruments 1405.

In this example, a fused process model 1810, of the fused dynamic state-space model 1805, includes one or more of a pulse oximeter physiology process model 1812, the hemodynamics process model 810, an electrocardiograph physiology model 1814, and/or the heart electrodynamics model 1110. For instance, the pulse oximeter physiology process model 1812 optionally incorporates one or more of the hemodynamics heart model 812, the hemodynamics vascular model 814, and/or the light scattering and/or absorbance model 816. Similarly, the electrocardiogram physiology process model 1814 optionally incorporates one or more of the heart electrodynamics model 1112 and/or the wave propagation model 1114.

In this example, a fused observation model 1820, of the fused dynamic state-space model 1805, includes one or more of a pulse oximeter observation noise model 1822, the hemodynamics observation model 820, an electrocardiograph noise model 1824, and/or the electrodynamics observation model 1120. For instance, the pulse oximeter observation noise model 1822 optionally incorporates one or more of the sensor dynamics and noise model 822 and the spectrometer signal transduction noise model 824. Similarly, the electrocardiograph observation noise model 1824 optionally incorporates one or more of the sensor noise and interference model 1122, the sensor dynamics model 1124, and/or the electrode placement model 1126. Any of the process model 1810 sub-models, such as the pulse oximeter physiology model 1812 and electrocardiogram physiology model 1814 share information or data with any of: another process model 1810 sub-model, the process model 1810, the observation model 1820, or any observation model 1820 sub-model, such as the pulse oximeter model 1822 and/or the electrocardiogram noise model 1824.

Generally, in a fused dynamic state-space model, the process model and observation model are optionally combined into a single model or are separate and share information. Further, any sub-model of the process model or sub-model of the observation model shares information or data with any other sub-model of the process model or observation model.

As described, supra, for the dynamic state-space model 210, the fused dynamic state-space model 1805 for the heart optionally receives inputs, including one or more of:

-   -   pulse oximeter and electrocardiograph device state parameters         1830;     -   pulse oximeter and electrocardiograph device model parameters         1840;     -   pulse oximeter and electrocardiograph device process noise         values 1850; and     -   pulse oximeter and electrocardiograph device observation noise         values 1860.

For example, the pulse oximeter and electrocardiograph device state parameters 1830 optionally include one or more of:

-   -   pulse oximeter related values of:         -   a radial pressure (P_(w));         -   an aortic pressure (P_(ao));         -   time (t);         -   a spectral intensity (I) or a related absorbance value;         -   a reflectance or reflectance ratio, such as a red             reflectance (R_(r)) or an infrared reflectance (R_(ir));             and/or         -   a spectral intensity ratio (I_(R)); and     -   electrocardiograph device related values of:         -   an atrium signal (AS); and/or         -   a ventricle signal (VS).

Example II

In another example, the electrocardiograph device observation parameters 1840 optionally include one or more of:

-   -   pulse oximeter related values of:         -   a heart rate (HR);         -   a stroke volume (SV); and/or         -   a whole-blood oxygen saturation (SpO₂); and     -   electrocardiograph device related values of:         -   a permittivity, (ε);         -   an autonomic nervous system (ANS) tone; and/or         -   a heart rate variability (HRV).             Fusion Benefits

Several non-limiting examples of the benefits of sensor fusion using at least one physiological model and a probabilistic processor 200 are provided.

Stroke Volume and Contractility

In a first case, fused, fusion, or fusing of sensor data from multiple instruments in combination with physical models of body systems yields additional information not present from any given instrument of the multiple instruments 1405. Without loss of generality, an example of generating a measure of stroke volume, a contractility, and/or a heart filling rate using data from a pulse oximeter and an electrocardiograph meter is used to demonstrate the indirect parameter estimation.

Herein, benefits of combining hemodynamic information with electrodynamic information in a fusion model is described. As described, supra, a pulse oximeter plethysmograph in combination with a hemodynamics physical model is used to determine a physical parameter not traditionally achieved from the pulse oximeter, such as a heartbeat stroke volume. Similarly, as described, supra, an electrocardiogram in combination with an electrodynamics physical model is used to determine a physical parameter not traditionally achieved from the electrocardiograph meter, such as contractility. Stroke volume and contractility are related, such as according to equation 34, SV≈FR·C  (34) where SV is stroke volume, FR, is the heart filing rate, and C is contractility. Here, the filling rate is determined using information indirectly measured by two systems (SV from the pulse oximeter and C from the ECG). Further, given a known or approximated filling rate, the electrocardiogram determined contractility gives information on the pulse oximeter determined stroke volume, and vise-versa.

In another case, fusing sensor data results in increased information for parameters determined with individual sensor data when the sensed data overlaps in terms of physiology and/or models thereof. For example, as stroke volume is an element of the heart model 812, which is tied to additional hemodynamic models in the hemodynamics dynamic state-space model, such as the vascular model 814, and the stroke volume is related to electrocardiograph data, as described supra, then the electrocardiograph signal optionally aids in determination of parameters directly or indirectly measured by the pulse oximeter and vise-versa. Generally, the electrodynamic signal is related to the hemodynamic signal through the use of one or more models, such as the hemodynamics dynamic state-space model 805, the electrocardiograph dynamic state-space model 1105, or a heart model combining two or more elements of the hemodynamics DSSM model 805 and the electrocardiograph DSSM model 1105.

Arrhythmia

As described, supra, in some systems, such as the heart, hemodynamic information and electrodynamic information are related. As described, supra, the hemodynamic information of stroke volume is related to the electrodynamic information of contractility. Hence, the hemodynamic information of the pulse oximeter yields additional information to any of the parameters measured by the electrocardiogram, such as an arrhythmia. Logically, if the heart is experiencing an arrhythmia, which is being detected by the electrocardiogram probabilistic model, then the heart is experiencing diminished stroke volume, as detected by the pulse oximeter. Hence, the hemodynamic information originating with the pulse oximeter provides supporting or combinatorial information to the electrocardiograph probabilistic model.

Similarly, a blood pressure meter yields information on blood pressure, which is related to heart function. Hence, blood pressure meter information is synergistic with electrocardiograph information and vise-versa. Further, blood pressure meter information is synergistic with hemodynamic, photoplethysmograph, and/or pulse oximeter information and vise-versa

Motion Artifact

In yet another example, patient movement results in a motion artifact in the sensed data of a given sensor. In many of the observation models 720 of the dynamic state-space model 210, a model is used that relates to sensor movement and/or movement of the body. As a first example, the hemodynamics dynamic state-space model 805 optionally uses the hemodynamics sensor dynamics and noise model 822. As a second example, the electrocardiogram dynamic state-space model 1105 optionally uses the sensor dynamic model 1124. Each of these models relate to movement of the sensor relative to the sensed element, such as the body. Hence, if the body moves, twitches, and/or experiences a bump in transport, such as in transport by an ambulance, the body movement may be detected as a motion artifact with a plurality of sensors. For example, the pulse oximeter and the electrocardiograph device may each detect the same motion artifact. Hence, fusion of the sensed data from multiple instruments allows the identification of an outlier signal or motion artifact signal in data from a first sensor through detection of the same motion artifact with a second sensor. Therefore, identification of a motion artifact with a first sensor is used to remove the same motion artifact from data from a second sensor. Optionally, an accelerometer is used to detect motion artifacts. The fusion of input sensor data from the accelerometer with data streams from one, two, or more additional devices allows removal of the motion artifact data from the one, two, or more additional devices.

Heart Rate Variability

In another example, sensor fusion is used to enhance a measure of heart rate variability. Generally, use of multiple sensors yields: (1) an over-determined system for outlier analysis and/or (2) varying sensor types where not all of the sensors are affected by a noise source. Herein, heart rate variability or variation in beat-to-beat interval of a heart is used to demonstrate each of these cases.

Heart rate variability is measured using a blood pressure meter, a photoplethysmograph derived from a pulse oximeter, or an electrocardiogram device. However, each of the blood pressure meter, pulse oximeter, and electrocardiogram device are subject to noise and/or patient motion artifacts, which result in false positive heartbeats and/or missed heartbeats.

Using a combination of sensors, such as the blood pressure meter, pulse oximeter, and/or electrocardiogram device, results in an over-determined system. The over-determined system allows for outlier analysis. By fusing the signals, an ambiguous signal from the first device is detected and overcome by use of the signal from the second measuring device.

Further, noise sources affecting a first measuring device, such as a pulse oximeter, are often separate from noise sources affecting a second measuring device, such as an electrocardiogram meter. For instance, electrical interference may affect an electrodynamic signal, such as the electrocardiograph, while not impacting a hemodynamic signal, such as a photoplethysmograph. By fusing the signals, noise is recognized in one sensor data stream at a given time as the noise source is not present in the second sensor data stream at the same time due to the noise source type not affecting both sensor types.

Environment Meter

In still yet another case, sensor output from one, two, or more instruments is additionally fused with output from an environmental meter. Herein, an environment meter senses one or more of: temperature, pressure, vibration, humidity, and/or position information, such as from a global positioning system. The environment meter information is used for outlier determination, error correction, calibration, and/or quality control or assurance.

Generally, fusion of signals or sensor data from a plurality of devices allows:

-   -   detection of a false positive or false negative signal from a         first device with a second device;     -   noise recognized in data from a first sensor type as the noise         is not present in a second sensor type;     -   fusion of environmental data with medical data;     -   determination of an additional parameter not measured or         independently measured with individual data types of the fused         data;     -   electrocardiograph data to aid in analysis of         photoplethysmograph data and vise-versa; and/or     -   electrodynamic information to aid in analysis of hemodynamic         information and vise-versa.         Hardware

The above description describes an apparatus for generation of a physiological estimate of a physiological process of an individual from input data, where the apparatus includes a biomedical monitoring device having a data processor configured to run a dual estimation algorithm, where the biomedical monitoring device is configured to produce the input data and where the input data includes at least one of: a photoplethysmogram and an electrocardiogram. The dual estimation algorithm is configured to use a dynamic state-space model to operate on the input data using both an iterative state estimator and an iterative model parameter estimator in generation of the physiological estimate, where the dynamic state-space model is configured to mathematically represent probabilities of physiological processes that generate the physiological estimate and mathematically represent probabilities of physical processes that affect collection of the input data. Generally, the algorithm is implemented using a data processor, such as in a computer, operable in or in conjunction with a biomedical monitoring device. The method and apparatus are optionally implemented in a rack system in a hospital intensive care unit, such as in connection, combination, and/or alongside other biomedical devices monitoring a patient and connected to a database system, alert station, monitoring station, recording system, nurse station, or a doctor interface.

More generally, the probabilistic digital signal processor is a physical processor, is integrated into a processor, such as in a computer, and/or is integrated into an analyzer. The analyzer is a physical device used to process data, such as sensor data 122. Optionally, the analyzer includes an input device, a power supply, a central processing unit, a memory storage unit, an output display screen, a communication port, and/or a wireless connector, such as Bluetooth. Preferably, the analyzer is integrated with a sensor, such as integrated into any of:

-   -   a pulse oximeter;     -   an electrocardiogram device;     -   a biomedical device;     -   a medical rack system;     -   a mechanical sensing system;     -   a complex machine;     -   a car;     -   a plane;     -   a fluid monitoring system; and/or     -   an oil transport line.

Optionally, the analyzer is configured to receive information from one or more sensors or instruments. Generally, the analyzer is configured for signal processing, filtering data, monitoring a parameter, generating a metric, estimating a parameter value, determining a parameter value, quality control, and/or quality assurance.

In another example, a cardiac stroke volume analyzer comprises a system processor, where the system processor comprises: (1) a probabilistic processor and (2) a dynamic state-space model. The cardiac stroke volume analyzer receives discrete first cardiovascular input data, related to a first sub-system of the biomedical system, from a first blood pressure instrument, such as a pulse oximeter, an electrocardiogram instrument, or a blood pressure analyzer, such as a blood pressure meter with a digital output operating on command, periodically, and/or in a semi-automated mode. The cardiac stroke volume analyzer receives discrete second cardiovascular input data, related to a second sub-system of the biomedical system, from a second electrocardiogram instrument, such as a pulse oximeter, an electrocardiogram instrument, or a blood pressure analyzer. Optionally, the cardiac stroke volume analyzer is an analyzer that, with or without stroke volume analysis, determines contractility or heart filling rate. Optionally and preferably, a system processor, of said cardiac stroke volume analyzer, fuses the first input data and the second input data into fused data, where the system processor comprises: (1) the probabilistic processor converting the fused data into at least two probability distribution functions and (2) at least one probabilistic model, of the dynamic state-space model, operating on the at least two probability distribution functions. Optionally and preferably, the system processor iteratively circulates at least two probability distribution functions in the dynamic state-space model in synchronization with receipt of at least one of: (1) updated first input data and (2) updated second input data. Generally, the system processor processes the probability distribution functions to generate an output related to the state of the biomedical system, such as a left ventricle stroke volume of a heart of a patient, a measure of contractility, and/or a measure of filling rate.

Additional Embodiments

In yet another embodiment, the method, system, and/or apparatus using a probabilistic model to extract physiological information from a biomedical sensor, described supra, optionally uses a sensor providing time-dependent signals. More particularly, pulse ox and ECG examples were provided, supra, to describe the use of the probabilistic model approach. However, the probabilistic model approach is more widely applicable.

The above description describes an apparatus for generation of a physiological estimate of a physiological process of an individual from input data, where the apparatus includes a biomedical monitoring device having a data processor configured to run a dual estimation algorithm, where the biomedical monitoring device is configured to produce the input data, and where the input data comprises at least one of: a photoplethysmogram and an electrocardiogram. The dual estimation algorithm is configured to use a dynamic state-space model to operate on the input data using both an iterative state estimator and an iterative model parameter estimator in generation of the physiological estimate, where the dynamic state-space model is configured to mathematically represent probabilities of physiological processes that generate the physiological estimate and mathematically represent probabilities of physical processes that affect collection of the input data. Generally, the algorithm is implemented using a data processor, such as in a computer, operable in or in conjunction with a biomedical monitoring device.

In yet another embodiment, the method, system, and/or apparatus using a probabilistic model to extract physiological information from a biomedical sensor, described supra, optionally uses a sensor providing time-dependent signals. More particularly, pulse ox and ECG examples were provided, infra, to describe the use of the probabilistic model approach. However, the probabilistic model approach is more widely applicable.

Some examples of physiological sensors used for input into the system with a corresponding physiological model include:

-   -   an ECG having about two to twelve leads yielding an ECG waveform         used to determine an RR-interval and/or various morphological         features related to arrhythmias;     -   pulse photoplethysmography yielding a PPG waveform for         determination of hemoglobins and/or total hemoglobin;     -   a multi-frequency PPG including multiple wavelengths to measure         a variety of gas concentration;     -   capnography or IR absorption yielding a real time waveform for         carbon dioxide determination, end-tidal CO₂, an inspired         minimum, and/or respiration rate;     -   a temperature sensor for continuous determination of core body         temperature and/or skin temperature;     -   an anesthetic gas sensor including nitrous oxide, N₂O, and         carbon dioxide, CO₂, used to determine minimum alveolar         concentration of an inhaled anesthetic;     -   a heart catheter yielding a thermodilution curve for         determination of a cardiac index and/or a blood temperature;     -   an impedance cardiography sensor yielding a thoracic electrical         bioimpedance reading for determination of thoracic fluid         content, accelerated cardiac index, stroke volume, cardiac         output, and/or systemic vascular resistance;     -   a mixed venous oxygen saturation catheter for determination of         SvO₂;     -   an electroencephalogram (EEG) yielding an EEG waveform and         characteristics thereof, such as spectral edge frequency, mean         dominant frequency, peak power frequency, compressed spectral         array analysis, color pattern display, and/or         delta-theta-alpha-beta band powers, any of which are used for         analysis of cardiac functions described herein;     -   electromyography (EMG) yielding an EMG waveform including         frequency measures, event detection, and/or amplitude of         contraction;     -   auscultation yielding sound pressure waveforms;     -   transcutaneous blood gas sensors for determination of carbon         dioxide, CO₂, and oxygen, O₂;     -   a pressure cuff yielding a pressure waveform for determination         of systolic pressure, diastolic pressure, mean arterial         pressure, heart rate, and/or hemodynamics;     -   spirometry combining capnography and flow waveforms for         information on respiratory rate, tidal volume, minute volume,         positive end-expiratory pressure, peak inspiratory pressure,         dynamic compliance, and/or airway resistance;     -   fetal and/or maternal sensors, such as ECG and sound         (auscultatory) sensors for determination of fetal movement,         heart rate, uterine activity, and/or maternal ECG;     -   laser Doppler flowmetry yielding a velocity waveform for         capillary blood flow rate;     -   an ultrasound and/or Doppler ultrasound yielding a waveform,         such as a two-dimensional or three-dimensional image, for         imaging and/or analysis of occlusion of blood vessel walls,         blood flow velocity profile, and/or other body site dependent         measures;     -   a perspirometer yielding a continuous or semi-continuous surface         impedance for information on skin perspiration levels; and/or     -   a digital medical history database to calibrate the model or to         screen the database for patient diseases and/or conditions.

Some examples of non-physiological sensors used for input into the system with a corresponding physiological model include:

-   -   an accelerometer;     -   a three axes accelerometer;     -   a gyroscope;     -   a compass;     -   light or a light reading;     -   a global positioning system, for air pressure data, ambient         light, humidity, and/or temperature;     -   a microphone; and/or     -   an ambient temperature sensor.

While specific dynamic state-space models and input and output parameters are provided for the purpose of describing the present method, the present invention is not limited to examples of the dynamic state-space models, sensors, biological monitoring devices, inputs, and/or outputs provided herein.

Diagnosis/Prognosis

Referring now to FIG. 19, the output of the probabilistic digital signal processor 200 optionally is used to diagnose 1910 a system element or component. The diagnosis 1910 is optionally used in a process of prognosis 1920 and/or in control 1930 of the system.

Integrated Blood Pressure Analyzer

In another embodiment of the invention, the blood pressure sensor is integrated into/with the cardiac stroke volume analyzer. Generally, an apparatus and/or method of use thereof is used for estimating state of a cardiovascular system of a person having a limb, comprising the steps of: (1) providing a cardiac stroke volume analyzer, comprising: (a) a blood pressure sensor, the blood pressure sensor generating a time-varying pressure state waveform output from a limb of the person; (b) a system processor connected to the blood pressure sensor; and (c) a dynamic state-space model; (2) the system processor receiving cardiovascular input data, from the blood pressure sensor, related to a transient pressure state of the cardiovascular system; (3) at least one probabilistic model, of the dynamic state-space model, operating on the time-varying pressure state waveform output to generate a probability distribution function to a non-pressure state of the cardiovascular system; (4) iteratively updating the probability distribution function using synchronized updated time-varying pressure state waveform output from the blood pressure sensor; and (5) the system processor processing the probability distribution function to generate a non-pressure state output related to at least one of a stroke volume of a heart of the person, central venous pressure, and arterial compliance of the person, the output displayed to at least one of the person and a doctor. Herein, central venous pressure is a measure of pressure in the superior vena cava, which can be used as an estimation of preload and right atrial pressure. Central venous pressure is often used as an assessment of hemodynamics in a patient, particularly in intensive care units. Traditionally, the central venous pressure can be measured using a central venous catheter placed in the superior vena cava near the right atrium. A normal central venous pressure reading is between 8 to 12 mmHg. This value can be changed depending on a patient's volume status or venous compliance.

In a first case, the blood pressure sensor comprises a blood pressure cuff, such as a traditional blood pressure cuff, coupled to a pressure transducer generating a pressure waveform output, where the pressure transducer is positioned between at least a portion of an inflatable blood pressure cuff circumferentially placed around a limb and the skin. Optionally and preferably, the blood pressure cuff is used to measure blood pressure using cuff pressures that are less than a systolic pressure, such as less than 95, 90, 80, 70, 60, 50, 40, or 30 percent of systolic pressure of a given individual, such as at pressures of less than 200, 175, 150, 140, 130, 120, 110, 100, 90, or 80 millimeters of mercury (mm Hg). Optionally and preferably at least two pressure waveforms are obtained, such as a first pressure waveform as a function of time at a first applied pressure, a second pressure waveform as a function of time at a second applied pressure, and/or an n^(th) pressure waveform as a function of time at an n^(th) pressure, where n comprises a positive integer of at least 1, 2, 3, 4, 5, or more. The n applied pressures differ from one another by at least 2, 4, 6, 8, 10, 20, 30, or 40 percent. Optionally, the lowest pressure is equal to a diastolic pressure and/or is within 10, 20, 30, or 40 percent of the diastolic pressure, where the systolic pressure and diastolic pressure for the individual are known, determined from a first measurement, use of a reference device, and/or are assumed to be in the range of systolic and diastolic pressures for a group of humans, such as all humans, Americans, males, females, in a given weight range, and/or for a given age. Optionally and preferably, the blood pressure cuff is an automated blood pressure cuff programmed to partially inflate and partially deflate in an alternating fashion at regular time intervals, on demand, or at irregular time periods, where the partial inflation pressures are any of the above described sub-peristaltic pressures and the partial deflation pressures are any of the above described pressures that are less than the most recent or any prior pressure for a given period of wearing the blood pressure cuff. Optionally, the blood pressure cuff is used to determine blood pressure of the individual using a non-inflated form of the blood pressure cuff, such as using passive pressure against the blood pressure transducer.

In a second case, an optical system is used to determine blood pressure, where the optical system uses 1, 2, 3, or more wavelengths of light, such as in a pulse oximeter. In the second case, the optical system is optionally integrated into the blood pressure sensor system. Optionally and preferably, the pulse oximeter generates an optical waveform as a function of time, for each of 1, 2, 3, or more wavelengths of light or combinations thereof, where the blood pressure system uses the photonic signal(s) as a function of time to determine the blood pressure. When using the pulse oximeter/optical waveform continuous input approach, calibrating the resultant blood pressure using an inflatable/deflatable blood pressure cuff, as in the first case described supra is optionally and preferably used.

In either case, the time-varying pressure waveform(s) are related to blood volume and then the blood volume is related to cardiovascular state, such as stroke volume, heart stroke volume, left ventricular stroke volume, arterial compliance, and/or aortic compliance. Any of the mathematical relationships described herein of the cardiovascular system are used to link the waveform outputs to the state parameters along with or separately with any mathematical description of a pressure, force, flow, or shape of an artery, capillary bed, vein, and/or heart component, such as using an equation with a first term related to a pressure to inflate an artery and a second term related to a second pressure radially stretch the artery.

The inventor/applicant notes that the method and apparatus for determination of a left ventricle stroke volume is deemed to be statutory subject matter under 35 U.S.C. § 101 as the method and apparatus, as claimed, is not a known technique and is certainly not: (1) routinely practiced in the art, (2) well-understood, (3) routine, (4) conventional, or (5) a basic building block of human knowledge. Further, the method and apparatus are not an implementation of a long standing, fundamental, and well-known practice. Particularly, the combination of additional elements, of: (1) a dynamic state-space model, (2) fusing sensor data, (3) a probabilistic updater, (4) iterative updating, and (5) the actual outcome of a measure not achievable by the individual medical device data, viewed in combination, amount to significantly more than the exception by meaningfully limiting the judicial exception.

Still yet another embodiment includes any combination and/or permutation of any of the elements described herein.

The particular implementations shown and described are illustrative of the invention and its best mode and are not intended to otherwise limit the scope of the present invention in any way. Indeed, for the sake of brevity, conventional manufacturing, connection, preparation, and other functional aspects of the system may not be described in detail. Furthermore, the connecting lines shown in the various figures are intended to represent exemplary functional relationships and/or physical couplings between the various elements. Many alternative or additional functional relationships or physical connections may be present in a practical system.

In the foregoing description, the invention has been described with reference to specific exemplary embodiments; however, it will be appreciated that various modifications and changes may be made without departing from the scope of the present invention as set forth herein. The description and figures are to be regarded in an illustrative manner, rather than a restrictive one and all such modifications are intended to be included within the scope of the present invention. Accordingly, the scope of the invention should be determined by the generic embodiments described herein and their legal equivalents rather than by merely the specific examples described above. For example, the steps recited in any method or process embodiment may be executed in any order and are not limited to the explicit order presented in the specific examples. Additionally, the components and/or elements recited in any apparatus embodiment may be assembled or otherwise operationally configured in a variety of permutations to produce substantially the same result as the present invention and are accordingly not limited to the specific configuration recited in the specific examples.

Benefits, other advantages and solutions to problems have been described above with regard to particular embodiments; however, any benefit, advantage, solution to problems or any element that may cause any particular benefit, advantage or solution to occur or to become more pronounced are not to be construed as critical, required or essential features or components.

As used herein, the terms “comprises”, “comprising”, or any variation thereof, are intended to reference a non-exclusive inclusion, such that a process, method, article, composition or apparatus that comprises a list of elements does not include only those elements recited, but may also include other elements not expressly listed or inherent to such process, method, article, composition or apparatus. Other combinations and/or modifications of the above-described structures, arrangements, applications, proportions, elements, materials or components used in the practice of the present invention, in addition to those not specifically recited, may be varied or otherwise particularly adapted to specific environments, manufacturing specifications, design parameters or other operating requirements without departing from the general principles of the same.

Although the invention has been described herein with reference to certain preferred embodiments, one skilled in the art will readily appreciate that other applications may be substituted for those set forth herein without departing from the spirit and scope of the present invention. Accordingly, the invention should only be limited by the Claims included below. 

The invention claimed is:
 1. A method for estimating a state of a cardiovascular system of a person having a limb, comprising the steps of: providing a cardiac stroke volume analyzer, comprising: a blood pressure sensor, said blood pressure sensor generating a time-varying pressure state waveform output from the limb of the person; a system processor connected to said blood pressure sensor; and a dynamic state-space model of a cardiovascular system; said system processor receiving cardiovascular input data, from said blood pressure sensor, related to a transient pressure state of the cardiovascular system; at least one probabilistic model, of said dynamic state-space model, operating on the time-varying pressure state waveform output to generate a probability distribution function to a non-pressure state of the cardiovascular system; iteratively updating said probability distribution function using synchronized updated time-varying pressure state waveform output from said blood pressure sensor; and said system processor processing the probability distribution function to generate a non-pressure state output related to an arterial compliance of the person, said output displayed on a display screen to at least one of the person and a medical professional.
 2. The method of claim 1, further comprising the steps of: providing an optical input to the limb of the person; and said blood pressure sensor generating a response to the optical input to yield said time-varying pressure state waveform output.
 3. The method of claim 2, further comprising the step of: relating the time-varying pressure state waveform output to a blood volume; and relating the blood volume to the arterial compliance.
 4. The method of claim 3, said step of relating further comprising the step of: using an equation with a first term related to a pressure to inflate an artery and a second term related to a second pressure to radially stretch the artery.
 5. The method of claim 1, further comprising the step of: a pressure transducer output measuring pressure from an alternating partial inflation of a blood pressure cuff and a partial deflation of said blood pressure cuff to yield said time-varying pressure state waveform output.
 6. The method of claim 1, further comprising the step of: a pressure transducer, of said blood pressure sensor, reading a passive change of the blood pressure of the person on a blood pressure cuff.
 7. The method of claim 1, said blood pressure sensor generating a time-varying pressure state waveform output further comprising the step of: using an automated blood pressure cuff to alternatingly measure systolic pressure and diastolic pressure at at least one of: a fixed time interval and a non-fixed time interval.
 8. The method of claim 1, further comprising the step of: measuring pressure directly as a function of time to yield said time varying pressure state waveform.
 9. The method of claim 1, further comprising the steps of: using a pulse oximeter to generate said time varying pressure state waveform; and using a blood pressure cuff to calibrate the time varying pressure state waveform.
 10. The method of claim 1, further comprising the step of: measuring pressure, using a blood pressure cuff alternatingly at a first partially inflated pressure in excess of 100 mm Hg and a second partially deflated pressure of less than 100 mm Hg, directly as a function of time to yield said time varying pressure state waveform.
 11. An apparatus for estimating state of a cardiovascular system of a person having a limb, comprising: a noninvasive cardiac stroke volume analyzer, comprising: a blood pressure sensor, said blood pressure sensor configured to generate a time-varying pressure state waveform output from the limb of the person, said cardiac stroke volume analyzer not comprising a Doppler sensor for measuring acoustic information; a system processor connected to said blood pressure sensor; and a dynamic state-space model of a cardiovascular system linked to said system processor; said system processor configured to receive cardiovascular input data, from said blood pressure sensor, related to a transient pressure state of the cardiovascular system; at least one probabilistic model, of said dynamic state-space model, configured to operate on the time-varying pressure state waveform output to generate a probability distribution function to a non-pressure state of the cardiovascular system; said system processor configured to iteratively update said probability distribution function using synchronized updated time-varying pressure state waveform output from said blood pressure sensor; and said system processor configured to process the probability distribution function to generate a non-pressure state output related to at least one of a stroke volume of a heart of the person and arterial compliance of the person, said output displayed to at least one of the person and a doctor.
 12. The apparatus of claim 11, said blood pressure sensor integrated into said cardiac stroke volume analyzer.
 13. The apparatus of claim 12, said blood pressure sensor comprising: a blood pressure cuff, said system processor configured to alternatingly partially inflate said blood pressure cuff to a sub-peristaltic pressure and to partially deflate said blood pressure cuff from said sub-peristaltic pressure to generate said time-varying pressure state waveform output.
 14. The apparatus of claim 13, further comprising: a computer display screen configured to display said output.
 15. The apparatus of claim 11, said blood pressure sensor comprising: an optical meter configured to provide an optical input to the limb of the person and responsive thereto generate said time-varying pressure state waveform; and a reference blood pressure cuff, said system processor configured to use output from said reference blood pressure cuff to calibrate said time-varying pressure state waveform.
 16. The apparatus of claim 15, said system processor configured to relate the time-varying pressure state waveform output to a blood volume and subsequently relate the blood volume to the arterial compliance.
 17. The apparatus of claim 11, said blood pressure sensor comprising a pressure transducer configured to output a pressure measure from an alternating partial inflation of a blood pressure cuff and a partial deflation of said blood pressure cuff to yield said time-varying pressure state waveform output.
 18. The apparatus of claim 11, said blood pressure sensor further comprising: a blood pressure cuff; and a pressure transducer between at least a portion of said blood pressure cuff and the limb of the person, said pressure transducer configured to generate and output a passive change of pressure of the person as the time-varying pressure state waveform output.
 19. The apparatus of claim 11, said blood pressure sensor further comprising: an automated blood pressure cuff configured to alternatingly measure systolic pressure and diastolic pressure at at least one of: (1) a fixed time interval and (2) a non-fixed time interval, and output the results as said time-varying pressure state waveform output.
 20. The apparatus of claim 19, said automated blood pressure cuff configured to measure pressure in the limb alternatingly at a first partially inflated and sub-peristaltic pressure and a second partially deflated pressure of less than said sub-peristaltic pressure directly as a function of time to yield said time varying pressure state waveform.
 21. An apparatus for estimating state of a cardiovascular system of a person having a limb, comprising: a noninvasive cardiac stroke volume analyzer, comprising: a blood pressure sensor, said blood pressure sensor configured to generate a time-varying pressure state waveform output from the limb of the person, said cardiac stroke volume analyzer not comprising an electrocardiogram sensor; a system processor connected to said blood pressure sensor; and a dynamic state-space model of a cardiovascular system linked to said system processor; said system processor configured to receive cardiovascular input data, from said blood pressure sensor, related to a transient pressure state of the cardiovascular system; at least one probabilistic model, of said dynamic state-space model, configured to operate on the time-varying pressure state waveform output to generate a probability distribution function to a non-pressure state of the cardiovascular system; said system processor configured to iteratively update said probability distribution function using synchronized updated time-varying pressure state waveform output from said blood pressure sensor; and said system processor configured to process the probability distribution function to generate a non-pressure state output related to at least one of a stroke volume of a heart of the person and arterial compliance of the person, said output displayed to at least one of the person and a doctor. 